-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 11:39:35 2024 UTC
-- File 20230906_171409_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.529549 to 19.490124
-each separately from 10s to 19s and from 489835us to 529838us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.959997 in 1 packets: 18
MIU packet time from 2023-09-06T17:14:22.529.549.000 to 2023-09-06T17:14:31.490.124.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.959997 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7092195 to 20.7091884
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999689 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_171409_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.2387684 to 12.2390574
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9599670  -- OK for SL --
   MIU time from 2023-09-06T17:14:22.529.549.000 to 2023-09-06T17:14:22.529.838.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.959997

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1990244 to 21.1993134
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T17:14:31.489.835.000 to 2023-09-06T17:14:31.490.124.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.238768400 corrected by 40bit 10MHz counter of 11.709219500 obtained at 1Hz counter of 10. Result: 10.529550700
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-06T17:14:22.529.408.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T17:14:31.489.983.725
   Min 1st sample XO time among all SL packets: 10.529410425
   Max 1st sample XO time among all SL packets: 19.489986391
   TM values from -262 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T17:14:22.529.408.725 (XO: 10.529410425 - 10.529716426, t=37.2 degC)  Rank: 00078
   TM values: Max =   220  Min =  -257  Ave =    -4.34  Std =    25.30
   Min TM value:-257 at an offset of 129.550 usec -> 2023-09-06T17:14:22.529.538.275 (XO: 10.529539975)
   Max TM value:220 at an offset of 117.288 usec -> 2023-09-06T17:14:22.529.526.012 (XO: 10.529527713)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 132.125 usec -> 2023-09-06T17:14:22.529.540.850 (XO: 10.529542550)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 117.425 usec -> 2023-09-06T17:14:22.529.526.150 (XO: 10.529527850)

 - Block number: 2 starting at 2023-09-06T17:14:31.489.694.725 (XO: 19.489697390 - 19.490003391, t=37.2 degC)  Rank: 00086
   TM values: Max =   225  Min =  -262  Ave =    -4.47  Std =    22.87
   Min TM value:-262 at an offset of 125.675 usec -> 2023-09-06T17:14:31.489.820.400 (XO: 19.489823065)
   Max TM value:225 at an offset of 113.125 usec -> 2023-09-06T17:14:31.489.807.850 (XO: 19.489810515)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 128.275 usec -> 2023-09-06T17:14:31.489.823.000 (XO: 19.489825665)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 113.263 usec -> 2023-09-06T17:14:31.489.807.987 (XO: 19.489810653)
