-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 11:14:01 2024 UTC
-- File 20230906_162636_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.815711 to 32.720457
-each separately from 11s to 32s and from 602680us to 816000us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.786680 in 1 packets: 18
      9.117199 in 1 packets: 36
MIU packet time from 2023-09-06T16:26:50.815.711.000 to 2023-09-06T16:27:11.720.457.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.786680 in 1 packets: 18
      9.117199 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.7029845 to 33.7029118
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999585 in 1 packets: 18
      8.9999688 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_162636_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.5186961 to 13.5189851
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7866390  -- OK for SL --
   MIU time from 2023-09-06T16:26:50.815.711.000 to 2023-09-06T16:26:50.816.000.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.786680

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.3056241 to 25.3059131
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1171670  -- OK for SL --
   MIU time from 2023-09-06T16:27:02.602.680.000 to 2023-09-06T16:27:02.602.969.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.117199

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4230801 to 34.4233691
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T16:27:11.720.168.000 to 2023-09-06T16:27:11.720.457.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.518696100 corrected by 40bit 10MHz counter of 12.702984500 obtained at 1Hz counter of 11. Result: 11.815714373
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-06T16:26:50.815.570.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T16:27:11.720.316.725
   Min 1st sample XO time among all SL packets: 11.815574098
   Max 1st sample XO time among all SL packets: 32.720319474
   TM values from -267 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T16:26:50.815.570.725 (XO: 11.815574098 - 11.815880099, t=37.2 degC)  Rank: 00080
   TM values: Max =   204  Min =  -267  Ave =    -4.34  Std =    22.04
   Min TM value:-267 at an offset of 120.725 usec -> 2023-09-06T16:26:50.815.691.450 (XO: 11.815694823)
   Max TM value:204 at an offset of 107.663 usec -> 2023-09-06T16:26:50.815.678.387 (XO: 11.815681761)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 118.363 usec -> 2023-09-06T16:26:50.815.689.087 (XO: 11.815692461)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 107.763 usec -> 2023-09-06T16:26:50.815.678.487 (XO: 11.815681861)

 - Block number: 2 starting at 2023-09-06T16:27:02.602.539.725 (XO: 23.602542934 - 23.602848935, t=37.2 degC)  Rank: 00079
   TM values: Max =   248  Min =  -257  Ave =    -4.43  Std =    20.57
   Min TM value:-257 at an offset of 121.188 usec -> 2023-09-06T16:27:02.602.660.912 (XO: 23.602664122)
   Max TM value:248 at an offset of 120.950 usec -> 2023-09-06T16:27:02.602.660.675 (XO: 23.602663884)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 118.725 usec -> 2023-09-06T16:27:02.602.658.450 (XO: 23.602661659)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 121.063 usec -> 2023-09-06T16:27:02.602.660.787 (XO: 23.602663997)

 - Block number: 3 starting at 2023-09-06T16:27:11.720.027.725 (XO: 32.720030473 - 32.720336474, t=37.2 degC)  Rank: 00083
   TM values: Max =   205  Min =  -238  Ave =    -4.46  Std =    15.84
   Min TM value:-238 at an offset of 123.650 usec -> 2023-09-06T16:27:11.720.151.375 (XO: 32.720154124)
   Max TM value:205 at an offset of 121.038 usec -> 2023-09-06T16:27:11.720.148.762 (XO: 32.720151511)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 123.750 usec -> 2023-09-06T16:27:11.720.151.475 (XO: 32.720154224)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 121.150 usec -> 2023-09-06T16:27:11.720.148.875 (XO: 32.720151624)
