-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 10:59:51 2024 UTC
-- File 20230906_161425_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.775719 to 20.355373
-each separately from 0s to 20s and from 355084us to 776008us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.579076 in 1 packets: 18
MIU packet time from 2023-09-06T16:14:28.775.719.000 to 2023-09-06T16:14:48.355.373.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.579076 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7049631 to 21.7048941
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999310 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_161425_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4806830 to 2.4809720
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5790070  -- OK for SL --
   MIU time from 2023-09-06T16:14:28.775.719.000 to 2023-09-06T16:14:28.776.008.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.579076

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0599790 to 22.0602680
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T16:14:48.355.084.000 to 2023-09-06T16:14:48.355.373.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.480683000 corrected by 40bit 10MHz counter of 1.704963100 obtained at 1Hz counter of 0. Result: 0.775722537
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-06T16:14:28.775.578.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T16:14:48.355.232.725
   Min 1st sample XO time among all SL packets: 0.775582262
   Max 1st sample XO time among all SL packets: 20.355234868
   TM values from -254 to 209 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T16:14:28.775.578.725 (XO: 0.775582262 - 0.775888263, t=37.2 degC)  Rank: 00079
   TM values: Max =   209  Min =  -254  Ave =    -4.46  Std =    23.26
   Min TM value:-254 at an offset of 98.413 usec -> 2023-09-06T16:14:28.775.677.137 (XO: 0.775680675)
   Max TM value:209 at an offset of 98.175 usec -> 2023-09-06T16:14:28.775.676.900 (XO: 0.775680437)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 114.838 usec -> 2023-09-06T16:14:28.775.693.562 (XO: 0.775697100)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 98.313 usec -> 2023-09-06T16:14:28.775.677.037 (XO: 0.775680575)

 - Block number: 2 starting at 2023-09-06T16:14:48.354.943.725 (XO: 20.354945867 - 20.355251868, t=37.2 degC)  Rank: 00081
   TM values: Max =   206  Min =  -233  Ave =    -4.39  Std =    19.27
   Min TM value:-233 at an offset of 127.725 usec -> 2023-09-06T16:14:48.355.071.450 (XO: 20.355073593)
   Max TM value:206 at an offset of 127.988 usec -> 2023-09-06T16:14:48.355.071.712 (XO: 20.355073855)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 127.850 usec -> 2023-09-06T16:14:48.355.071.575 (XO: 20.355073718)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 125.200 usec -> 2023-09-06T16:14:48.355.068.925 (XO: 20.355071068)
