-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 21:31:14 2024 UTC
-- File 20230722_085135_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.767594 to 15.818396
-each separately from 5s to 15s and from 767594us to 818396us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.050224 in 1 packets: 18
MIU packet time from 2023-07-22T08:51:43.767.594.000 to 2023-07-22T08:51:53.818.396.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.050224 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7143283 to 16.7142954
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999671 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230722_085135_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      126 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4819246 to 7.4822136
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0501910  -- OK for SL --
   MIU time from 2023-07-22T08:51:43.767.594.000 to 2023-07-22T08:51:43.767.883.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.050224

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.5324046 to 17.5326936
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-22T08:51:53.818.107.000 to 2023-07-22T08:51:53.818.396.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.481924600 corrected by 40bit 10MHz counter of 6.714328300 obtained at 1Hz counter of 5. Result: 5.767598833
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-07-22T08:51:43.767.453.725
   Max 1st sample MIU time among all SL packets: 2023-07-22T08:51:53.818.255.725
   Min 1st sample XO time among all SL packets: 5.767458558
   Max 1st sample XO time among all SL packets: 15.818260625
   TM values from -941 to 1066 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-22T08:51:43.767.453.725 (XO: 5.767458558 - 5.767764559, t=45.2 degC)  Rank: 00126
   TM values: Max =  1066  Min =  -941  Ave =    -4.51  Std =    45.09
   Min TM value:-941 at an offset of 90.175 usec -> 2023-07-22T08:51:43.767.543.900 (XO: 5.767548733)
   Max TM value:1066 at an offset of 89.913 usec -> 2023-07-22T08:51:43.767.543.637 (XO: 5.767548470)
   Max dTM/dt [1/us]:1.32e+04 at an offset of 89.850 usec -> 2023-07-22T08:51:43.767.543.575 (XO: 5.767548408)
   Min dTM/dt [1/us]:-1.29e+04 at an offset of 89.963 usec -> 2023-07-22T08:51:43.767.543.687 (XO: 5.767548520)

 - Block number: 2 starting at 2023-07-22T08:51:53.817.966.725 (XO: 15.817971624 - 15.818277625, t=45.2 degC)  Rank: 00080
   TM values: Max =   200  Min =  -305  Ave =    -4.49  Std =    26.42
   Min TM value:-305 at an offset of 115.813 usec -> 2023-07-22T08:51:53.818.082.537 (XO: 15.818087437)
   Max TM value:200 at an offset of 116.100 usec -> 2023-07-22T08:51:53.818.082.825 (XO: 15.818087725)
   Max dTM/dt [1/us]:2.50e+03 at an offset of 115.925 usec -> 2023-07-22T08:51:53.818.082.650 (XO: 15.818087550)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 113.463 usec -> 2023-07-22T08:51:53.818.080.187 (XO: 15.818085087)
