-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan  8 21:03:06 2024 UTC
-- File 20230722_063641_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177025
-each separately from 0s to 14s and from 0us to 841251us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU packet time from 2023-07-22T06:36:44.000.000.000 to 2023-07-22T06:36:58.177.025.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5870956 to 15.5870496
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999967 in 10 packets: 2, 3, 4, 5, 6, 9, 10, 15, 16, 17
      0.9999968 in 3 packets: 8, 11, 14
      0.9999966 in 1 packets: 12
   XO PPM: 
      -3.3 in 10 packets: 2, 3, 4, 5, 6, 9, 10, 15, 16, 17
      -3.2 in 3 packets: 8, 11, 14
      -3.4 in 1 packets: 12
Temperature FPGA: 
      00 -> 55.0 degC in 9 packets: 1, 2, 4, 5, 6, 7, 8, 12, 16
      FF -> 56.3 degC in 9 packets: 3, 9, 10, 11, 13, 14, 15, 17, 18
Temperature XO: 
      C6 -> 45.2 degC in 8 packets: 1, 2, 4, 10, 12, 13, 14, 15
      C5 -> 43.6 degC in 10 packets: 3, 5, 6, 7, 8, 9, 11, 16, 17, 18
Temperature ant 3: 
      02 -> 100.7 degC in 3 packets: 1, 2, 3
      01 -> 108.0 degC in 15 packets: 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      1 in 1 packets: 17
      2 in 1 packets: 18
Number of generated alerts: 
      0 in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      1 in 1 packets: 17
      2 in 1 packets: 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230722_063641_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5870957 to 15.7640757
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-22T06:36:44.000.000.000 to 2023-07-22T06:36:58.177.025.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.587095700 corrected by 40bit 10MHz counter of 1.587095600 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SR packets: 2023-07-22T06:36:43.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-22T06:36:58.177.024.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177026609
   TM values from -22 to 13 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-22T06:36:43.999.999.925 (XO: 0.000000025 - 0.000017025, t=45.2 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.48  Std =     6.38

 - Packet number: 2 starting at 2023-07-22T06:36:44.833.937.925 (XO: 0.833942777 - 0.833959777, t=45.2 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.36  Std =     7.24

 - Packet number: 3 starting at 2023-07-22T06:36:45.667.880.925 (XO: 1.667885529 - 1.667902529, t=43.6 degC)
   TM values: Max =    10  Min =   -17  Ave =    -4.34  Std =     5.61

 - Packet number: 4 starting at 2023-07-22T06:36:46.501.824.925 (XO: 2.501828281 - 2.501845281, t=45.2 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.46  Std =     7.39

 - Packet number: 5 starting at 2023-07-22T06:36:47.335.767.925 (XO: 3.335771033 - 3.335788033, t=43.6 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.57  Std =     3.43

 - Packet number: 6 starting at 2023-07-22T06:36:48.169.711.925 (XO: 4.169713785 - 4.169730785, t=43.6 degC)
   TM values: Max =     8  Min =   -17  Ave =    -4.37  Std =     4.58

 - Packet number: 7 starting at 2023-07-22T06:36:49.003.654.925 (XO: 5.003656537 - 5.003673537, t=43.6 degC)
   TM values: Max =     9  Min =   -19  Ave =    -4.60  Std =     5.78

 - Packet number: 8 starting at 2023-07-22T06:36:49.837.593.925 (XO: 5.837599205 - 5.837616205, t=43.6 degC)
   TM values: Max =     8  Min =   -20  Ave =    -4.68  Std =     5.56

 - Packet number: 9 starting at 2023-07-22T06:36:50.671.537.925 (XO: 6.671541941 - 6.671558941, t=43.6 degC)
   TM values: Max =     6  Min =   -13  Ave =    -4.36  Std =     3.06

 - Packet number: 10 starting at 2023-07-22T06:36:51.505.480.925 (XO: 7.505484693 - 7.505501693, t=45.2 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.72  Std =     5.89

 - Packet number: 11 starting at 2023-07-22T06:36:52.339.424.925 (XO: 8.339427411 - 8.339444411, t=43.6 degC)
   TM values: Max =     6  Min =   -13  Ave =    -4.33  Std =     3.71

 - Packet number: 12 starting at 2023-07-22T06:36:53.173.367.925 (XO: 9.173370114 - 9.173387115, t=45.2 degC)
   TM values: Max =     6  Min =   -16  Ave =    -4.41  Std =     4.52

 - Packet number: 13 starting at 2023-07-22T06:36:54.007.311.925 (XO: 10.007312948 - 10.007329948, t=45.2 degC)
   TM values: Max =     9  Min =   -17  Ave =    -4.57  Std =     4.08

 - Packet number: 14 starting at 2023-07-22T06:36:54.841.250.925 (XO: 10.841255617 - 10.841272617, t=45.2 degC)
   TM values: Max =     9  Min =   -17  Ave =    -4.69  Std =     4.98

 - Packet number: 15 starting at 2023-07-22T06:36:55.675.193.925 (XO: 11.675198353 - 11.675215353, t=45.2 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.33  Std =     2.25

 - Packet number: 16 starting at 2023-07-22T06:36:56.509.137.925 (XO: 12.509141105 - 12.509158105, t=43.6 degC)
   TM values: Max =    12  Min =   -20  Ave =    -4.76  Std =     8.33

 - Packet number: 17 starting at 2023-07-22T06:36:57.343.080.925 (XO: 13.343083857 - 13.343100857, t=43.6 degC)
   TM values: Max =    12  Min =   -21  Ave =    -4.57  Std =     6.70

 - Packet number: 18 starting at 2023-07-22T06:36:58.177.024.925 (XO: 14.177026609 - 14.177043609, t=43.6 degC)
   TM values: Max =     9  Min =   -17  Ave =    -4.57  Std =     6.01
