-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul  7 04:51:36 2023 UTC
-- File 20230628_094312_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177028
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2023-06-28T09:43:15.000.000.000 to 2023-06-28T09:43:29.177.028.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9050582 to 15.9050093
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 9 packets: 2, 3, 4, 5, 6, 8, 9, 10, 11
      0.9999966 in 3 packets: 12, 15, 17
      0.9999964 in 2 packets: 14, 16
   XO PPM: 
      -3.5 in 9 packets: 2, 3, 4, 5, 6, 8, 9, 10, 11
      -3.4 in 3 packets: 12, 15, 17
      -3.6 in 2 packets: 14, 16
Temperature FPGA: 
      03 -> 51.1 degC in 2 packets: 1, 10
      01 -> 53.7 degC in 5 packets: 2, 3, 4, 6, 9
      02 -> 52.4 degC in 11 packets: 5, 7, 8, 11, 12, 13, 14, 15, 16, 17, 18
Temperature XO: 
      C3 -> 40.4 degC in 13 packets: 1, 4, 5, 7, 8, 9, 11, 12, 13, 14, 16, 17, 18
      C4 -> 42.0 degC in 5 packets: 2, 3, 6, 10, 15
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 12 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12
      1 in 3 packets: 13, 14, 15
      2 in 3 packets: 16, 17, 18
Number of generated alerts: 
      0 in 12 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12
      1 in 3 packets: 13, 14, 15
      2 in 3 packets: 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230628_094312_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9050583 to 16.0820383
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-28T09:43:15.000.000.000 to 2023-06-28T09:43:29.177.028.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 11, 14, 16
      0.833944 in 7 packets: 3, 5, 8, 10, 12, 15, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.905058300 corrected by 40bit 10MHz counter of 1.905058200 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996600
   Min 1st sample MIU time among all SR packets: 2023-06-28T09:43:14.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-06-28T09:43:29.177.027.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177029527
   TM values from -14 to 5 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-06-28T09:43:14.999.999.925 (XO: 0.000000025 - 0.000017025, t=40.4 degC)
   TM values: Max =     1  Min =   -11  Ave =    -4.60  Std =     1.70

 - Packet number: 2 starting at 2023-06-28T09:43:15.833.938.925 (XO: 0.833942944 - 0.833959944, t=42.0 degC)
   TM values: Max =     1  Min =   -10  Ave =    -4.34  Std =     1.64

 - Packet number: 3 starting at 2023-06-28T09:43:16.667.881.925 (XO: 1.667885863 - 1.667902863, t=42.0 degC)
   TM values: Max =     1  Min =   -12  Ave =    -4.35  Std =     1.68

 - Packet number: 4 starting at 2023-06-28T09:43:17.501.825.925 (XO: 2.501828781 - 2.501845781, t=40.4 degC)
   TM values: Max =     5  Min =   -13  Ave =    -4.51  Std =     1.76

 - Packet number: 5 starting at 2023-06-28T09:43:18.335.768.925 (XO: 3.335771700 - 3.335788700, t=40.4 degC)
   TM values: Max =     3  Min =   -12  Ave =    -4.33  Std =     1.94

 - Packet number: 6 starting at 2023-06-28T09:43:19.169.712.925 (XO: 4.169714619 - 4.169731619, t=42.0 degC)
   TM values: Max =     1  Min =   -12  Ave =    -4.47  Std =     1.90

 - Packet number: 7 starting at 2023-06-28T09:43:20.003.655.925 (XO: 5.003657538 - 5.003674538, t=40.4 degC)
   TM values: Max =     1  Min =   -10  Ave =    -4.77  Std =     1.72

 - Packet number: 8 starting at 2023-06-28T09:43:20.837.595.925 (XO: 5.837600457 - 5.837617457, t=40.4 degC)
   TM values: Max =     5  Min =   -12  Ave =    -4.32  Std =     1.96

 - Packet number: 9 starting at 2023-06-28T09:43:21.671.539.925 (XO: 6.671543375 - 6.671560375, t=40.4 degC)
   TM values: Max =     3  Min =   -12  Ave =    -4.36  Std =     1.90

 - Packet number: 10 starting at 2023-06-28T09:43:22.505.482.925 (XO: 7.505486294 - 7.505503294, t=42.0 degC)
   TM values: Max =     4  Min =   -12  Ave =    -4.55  Std =     1.98

 - Packet number: 11 starting at 2023-06-28T09:43:23.339.426.925 (XO: 8.339429213 - 8.339446213, t=40.4 degC)
   TM values: Max =     2  Min =   -11  Ave =    -4.43  Std =     1.66

 - Packet number: 12 starting at 2023-06-28T09:43:24.173.369.925 (XO: 9.173372114 - 9.173389115, t=40.4 degC)
   TM values: Max =     1  Min =   -10  Ave =    -4.48  Std =     1.69

 - Packet number: 13 starting at 2023-06-28T09:43:25.007.313.925 (XO: 10.007314951 - 10.007331951, t=40.4 degC)
   TM values: Max =     2  Min =   -12  Ave =    -4.46  Std =     1.84

 - Packet number: 14 starting at 2023-06-28T09:43:25.841.253.925 (XO: 10.841257954 - 10.841274954, t=40.4 degC)
   TM values: Max =     2  Min =   -11  Ave =    -4.38  Std =     1.82

 - Packet number: 15 starting at 2023-06-28T09:43:26.675.196.925 (XO: 11.675200821 - 11.675217821, t=42.0 degC)
   TM values: Max =     4  Min =   -12  Ave =    -4.63  Std =     1.86

 - Packet number: 16 starting at 2023-06-28T09:43:27.509.140.925 (XO: 12.509143758 - 12.509160758, t=40.4 degC)
   TM values: Max =     0  Min =    -9  Ave =    -4.39  Std =     1.59

 - Packet number: 17 starting at 2023-06-28T09:43:28.343.083.925 (XO: 13.343086691 - 13.343103692, t=40.4 degC)
   TM values: Max =     0  Min =    -9  Ave =    -4.41  Std =     1.66

 - Packet number: 18 starting at 2023-06-28T09:43:29.177.027.925 (XO: 14.177029527 - 14.177046527, t=40.4 degC)
   TM values: Max =     5  Min =   -14  Ave =    -4.45  Std =     1.79
