-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul  7 03:41:58 2023 UTC
-- File 20230628_081300_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.616668 to 14.616957
-each separately from 14s to 14s and from 616668us to 616957us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2023-06-28T08:13:17.616.668.000 to 2023-06-28T08:13:17.616.957.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7407076 to 15.7407076
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      FF -> 56.3 degC in 18 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230628_081300_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.3573776 to 16.3576666
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-28T08:13:17.616.668.000 to 2023-06-28T08:13:17.616.957.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.357377600 corrected by 40bit 10MHz counter of 14.740711100 obtained at 1Hz counter of 13. Result: 14.616672158
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-28T08:13:17.616.527.725
   Max 1st sample MIU time among all SL packets: 2023-06-28T08:13:17.616.816.725
   Min 1st sample XO time among all SL packets: 14.616531883
   Max 1st sample XO time among all SL packets: 14.616820884
   TM values from -269 to 199 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-28T08:13:17.616.527.725 (XO: 14.616531883 - 14.616837884, t=45.2 degC)  Rank: 00078
   TM values: Max =   199  Min =  -269  Ave =    -4.29  Std =    24.04
   Min TM value:-269 at an offset of 97.338 usec -> 2023-06-28T08:13:17.616.625.062 (XO: 14.616629221)
   Max TM value:199 at an offset of 97.125 usec -> 2023-06-28T08:13:17.616.624.850 (XO: 14.616629008)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 99.750 usec -> 2023-06-28T08:13:17.616.627.475 (XO: 14.616631633)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 97.225 usec -> 2023-06-28T08:13:17.616.624.950 (XO: 14.616629108)
