-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul  7 01:56:29 2023 UTC
-- File 20230628_045445_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.596203 to 15.556703
-each separately from 0s to 15s and from 556414us to 596492us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.959922 in 1 packets: 18
MIU packet time from 2023-06-28T04:54:48.596.203.000 to 2023-06-28T04:55:03.556.703.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.959922 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7083980 to 16.7083471
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999491 in 1 packets: 18
Temperature FPGA: 
      FE -> 57.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230628_045445_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3046028 to 2.3048918
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9598710  -- OK for SL --
   MIU time from 2023-06-28T04:54:48.596.203.000 to 2023-06-28T04:54:48.596.492.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.959922

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.2647628 to 17.2650518
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-28T04:55:03.556.414.000 to 2023-06-28T04:55:03.556.703.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.304602800 corrected by 40bit 10MHz counter of 1.708398000 obtained at 1Hz counter of 0. Result: 0.596206887
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-28T04:54:48.596.062.725
   Max 1st sample MIU time among all SL packets: 2023-06-28T04:55:03.556.562.725
   Min 1st sample XO time among all SL packets: 0.596066611
   Max 1st sample XO time among all SL packets: 15.556566317
   TM values from -302 to 205 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-28T04:54:48.596.062.725 (XO: 0.596066611 - 0.596372612, t=45.2 degC)  Rank: 00083
   TM values: Max =   204  Min =  -279  Ave =    -4.39  Std =    24.14
   Min TM value:-279 at an offset of 129.150 usec -> 2023-06-28T04:54:48.596.191.875 (XO: 0.596195762)
   Max TM value:204 at an offset of 128.900 usec -> 2023-06-28T04:54:48.596.191.625 (XO: 0.596195512)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 131.600 usec -> 2023-06-28T04:54:48.596.194.325 (XO: 0.596198212)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 129.063 usec -> 2023-06-28T04:54:48.596.191.787 (XO: 0.596195674)

 - Block number: 2 starting at 2023-06-28T04:55:03.556.273.725 (XO: 15.556277316 - 15.556583317, t=45.2 degC)  Rank: 00077
   TM values: Max =   205  Min =  -302  Ave =    -4.41  Std =    23.30
   Min TM value:-302 at an offset of 93.338 usec -> 2023-06-28T04:55:03.556.367.062 (XO: 15.556370654)
   Max TM value:205 at an offset of 95.850 usec -> 2023-06-28T04:55:03.556.369.575 (XO: 15.556373167)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 95.688 usec -> 2023-06-28T04:55:03.556.369.412 (XO: 15.556373004)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 93.188 usec -> 2023-06-28T04:55:03.556.366.912 (XO: 15.556370504)
