-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul  7 01:45:12 2023 UTC
-- File 20230628_044033_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.230440 to 12.230729
-each separately from 12s to 12s and from 230440us to 230729us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2023-06-28T04:40:48.230.440.000 to 2023-06-28T04:40:48.230.729.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.2260734 to 14.2260734
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      FE -> 57.6 degC in 18 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230628_044033_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4565144 to 14.4568034
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-28T04:40:48.230.440.000 to 2023-06-28T04:40:48.230.729.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.456514400 corrected by 40bit 10MHz counter of 14.226073400 obtained at 1Hz counter of 12. Result: 12.230441784
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-28T04:40:48.230.299.725
   Max 1st sample MIU time among all SL packets: 2023-06-28T04:40:48.230.588.725
   Min 1st sample XO time among all SL packets: 12.230301508
   Max 1st sample XO time among all SL packets: 12.230590509
   TM values from -272 to 233 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-28T04:40:48.230.299.725 (XO: 12.230301508 - 12.230607509, t=45.2 degC)  Rank: 00075
   TM values: Max =   233  Min =  -272  Ave =    -4.42  Std =    23.57
   Min TM value:-272 at an offset of 116.025 usec -> 2023-06-28T04:40:48.230.415.750 (XO: 12.230417533)
   Max TM value:233 at an offset of 115.775 usec -> 2023-06-28T04:40:48.230.415.500 (XO: 12.230417283)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 118.375 usec -> 2023-06-28T04:40:48.230.418.100 (XO: 12.230419883)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 115.888 usec -> 2023-06-28T04:40:48.230.415.612 (XO: 12.230417396)
