-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul  7 01:38:46 2023 UTC
-- File 20230628_043223_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.582260 to 21.932613
-each separately from 2s to 21s and from 582260us to 932613us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.349775 in 1 packets: 18
MIU packet time from 2023-06-28T04:32:28.582.260.000 to 2023-06-28T04:32:47.932.613.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.349775 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7063545 to 22.7062903
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999358 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230628_043223_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2886162 to 4.2889052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3497110  -- OK for SL --
   MIU time from 2023-06-28T04:32:28.582.260.000 to 2023-06-28T04:32:28.582.549.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.349775

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6386162 to 23.6389052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-28T04:32:47.932.324.000 to 2023-06-28T04:32:47.932.613.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.288616200 corrected by 40bit 10MHz counter of 3.706354500 obtained at 1Hz counter of 2. Result: 2.582263680
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-28T04:32:28.582.119.725
   Max 1st sample MIU time among all SL packets: 2023-06-28T04:32:47.932.472.725
   Min 1st sample XO time among all SL packets: 2.582123404
   Max 1st sample XO time among all SL packets: 21.932477702
   TM values from -261 to 214 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-28T04:32:28.582.119.725 (XO: 2.582123404 - 2.582429405, t=45.2 degC)  Rank: 00090
   TM values: Max =   214  Min =  -261  Ave =    -4.37  Std =    29.50
   Min TM value:-261 at an offset of 110.900 usec -> 2023-06-28T04:32:28.582.230.625 (XO: 2.582234305)
   Max TM value:214 at an offset of 115.850 usec -> 2023-06-28T04:32:28.582.235.575 (XO: 2.582239255)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 113.375 usec -> 2023-06-28T04:32:28.582.233.100 (XO: 2.582236780)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 110.738 usec -> 2023-06-28T04:32:28.582.230.462 (XO: 2.582234142)

 - Block number: 2 starting at 2023-06-28T04:32:47.932.183.725 (XO: 21.932188701 - 21.932494702, t=45.2 degC)  Rank: 00078
   TM values: Max =   188  Min =  -254  Ave =    -4.40  Std =    24.21
   Min TM value:-254 at an offset of 125.275 usec -> 2023-06-28T04:32:47.932.309.000 (XO: 21.932313977)
   Max TM value:188 at an offset of 128.075 usec -> 2023-06-28T04:32:47.932.311.800 (XO: 21.932316777)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 125.425 usec -> 2023-06-28T04:32:47.932.309.150 (XO: 21.932314127)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 111.588 usec -> 2023-06-28T04:32:47.932.295.312 (XO: 21.932300289)
