-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul  7 01:34:01 2023 UTC
-- File 20230628_042800_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.439067 to 26.579599
-each separately from 6s to 26s and from 439067us to 579599us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.139954 in 1 packets: 18
MIU packet time from 2023-06-28T04:28:09.439.067.000 to 2023-06-28T04:28:29.579.599.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.139954 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.7072163 to 27.7071488
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999325 in 1 packets: 18
Temperature FPGA: 
      FE -> 57.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FF -> 56.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230628_042800_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.1462849 to 8.1465739
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1398870  -- OK for SL --
   MIU time from 2023-06-28T04:28:09.439.067.000 to 2023-06-28T04:28:09.439.356.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.139954

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.2864609 to 28.2867499
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-28T04:28:29.579.310.000 to 2023-06-28T04:28:29.579.599.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.146284900 corrected by 40bit 10MHz counter of 7.707216300 obtained at 1Hz counter of 6. Result: 6.439070093
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-28T04:28:09.438.926.725
   Max 1st sample MIU time among all SL packets: 2023-06-28T04:28:29.579.458.725
   Min 1st sample XO time among all SL packets: 6.438929817
   Max 1st sample XO time among all SL packets: 26.579462795
   TM values from -293 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-28T04:28:09.438.926.725 (XO: 6.438929817 - 6.439235818, t=45.2 degC)  Rank: 00081
   TM values: Max =   226  Min =  -290  Ave =    -4.32  Std =    23.07
   Min TM value:-290 at an offset of 118.875 usec -> 2023-06-28T04:28:09.439.045.600 (XO: 6.439048693)
   Max TM value:226 at an offset of 118.625 usec -> 2023-06-28T04:28:09.439.045.350 (XO: 6.439048443)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 121.213 usec -> 2023-06-28T04:28:09.439.047.937 (XO: 6.439051030)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 118.738 usec -> 2023-06-28T04:28:09.439.045.462 (XO: 6.439048555)

 - Block number: 2 starting at 2023-06-28T04:28:29.579.169.725 (XO: 26.579173794 - 26.579479795, t=45.2 degC)  Rank: 00083
   TM values: Max =   217  Min =  -293  Ave =    -4.30  Std =    21.75
   Min TM value:-293 at an offset of 81.863 usec -> 2023-06-28T04:28:29.579.251.587 (XO: 26.579255657)
   Max TM value:217 at an offset of 81.625 usec -> 2023-06-28T04:28:29.579.251.350 (XO: 26.579255419)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 84.213 usec -> 2023-06-28T04:28:29.579.253.937 (XO: 26.579258007)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 81.750 usec -> 2023-06-28T04:28:29.579.251.475 (XO: 26.579255544)
