-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jul  5 01:31:59 2023 UTC
-- File 20230626_211409_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.933500 to 17.363618
-each separately from 11s to 17s and from 363329us to 933789us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.429540 in 1 packets: 18
MIU packet time from 2023-06-26T21:14:23.933.500.000 to 2023-06-26T21:14:29.363.618.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.429540 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.7099816 to 18.7099615
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999799 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230626_211409_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.6434843 to 13.6437733
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.4295190  -- OK for SL --
   MIU time from 2023-06-26T21:14:23.933.500.000 to 2023-06-26T21:14:23.933.789.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.429540

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.0732923 to 19.0735813
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-26T21:14:29.363.329.000 to 2023-06-26T21:14:29.363.618.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.643484300 corrected by 40bit 10MHz counter of 12.709981600 obtained at 1Hz counter of 11. Result: 11.933505781
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-26T21:14:23.933.359.725
   Max 1st sample MIU time among all SL packets: 2023-06-26T21:14:29.363.477.725
   Min 1st sample XO time among all SL packets: 11.933365505
   Max 1st sample XO time among all SL packets: 17.363480724
   TM values from -327 to 245 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-26T21:14:23.933.359.725 (XO: 11.933365505 - 11.933671506, t=45.2 degC)  Rank: 00084
   TM values: Max =   245  Min =  -327  Ave =    -4.49  Std =    30.63
   Min TM value:-327 at an offset of 82.850 usec -> 2023-06-26T21:14:23.933.442.575 (XO: 11.933448355)
   Max TM value:245 at an offset of 87.688 usec -> 2023-06-26T21:14:23.933.447.412 (XO: 11.933453193)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 85.200 usec -> 2023-06-26T21:14:23.933.444.925 (XO: 11.933450705)
   Min dTM/dt [1/us]:-2.60e+03 at an offset of 82.688 usec -> 2023-06-26T21:14:23.933.442.412 (XO: 11.933448193)

 - Block number: 2 starting at 2023-06-26T21:14:29.363.188.725 (XO: 17.363191724 - 17.363497725, t=45.2 degC)  Rank: 00084
   TM values: Max =   224  Min =  -277  Ave =    -4.47  Std =    32.91
   Min TM value:-277 at an offset of 124.288 usec -> 2023-06-26T21:14:29.363.313.012 (XO: 17.363316011)
   Max TM value:224 at an offset of 148.463 usec -> 2023-06-26T21:14:29.363.337.187 (XO: 17.363340187)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 148.400 usec -> 2023-06-26T21:14:29.363.337.125 (XO: 17.363340124)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 148.600 usec -> 2023-06-26T21:14:29.363.337.325 (XO: 17.363340324)
