-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jul  5 01:29:03 2023 UTC
-- File 20230626_211031_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.147230 to 45.257135
-each separately from 2s to 45s and from 147230us to 917306us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.769498 in 1 packets: 18
      10.959684 in 1 packets: 36
      13.379567 in 1 packets: 54
MIU packet time from 2023-06-26T21:10:36.147.230.000 to 2023-06-26T21:11:19.257.135.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.769498 in 1 packets: 18
      10.959684 in 1 packets: 36
      13.379567 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.6970590 to 46.6969147
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999396 in 1 packets: 18
      10.9999631 in 1 packets: 36
      13.9999530 in 1 packets: 54
Temperature FPGA: 
      FE -> 57.6 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FF -> 56.3 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C6 -> 45.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230626_211031_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.8442901 to 3.8445791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7694390  -- OK for SL --
   MIU time from 2023-06-26T21:10:36.147.230.000 to 2023-06-26T21:10:36.147.519.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.769498

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6140181 to 22.6143071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9596470  -- OK for SL --
   MIU time from 2023-06-26T21:10:54.917.017.000 to 2023-06-26T21:10:54.917.306.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.959684

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.5739541 to 33.5742431
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3795190  -- OK for SL --
   MIU time from 2023-06-26T21:11:05.876.990.000 to 2023-06-26T21:11:05.877.279.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.379567

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.9537621 to 46.9540511
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-26T21:11:19.256.846.000 to 2023-06-26T21:11:19.257.135.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.844290100 corrected by 40bit 10MHz counter of 3.697059000 obtained at 1Hz counter of 2. Result: 2.147231601
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-26T21:10:36.147.089.725
   Max 1st sample MIU time among all SL packets: 2023-06-26T21:11:19.256.994.725
   Min 1st sample XO time among all SL packets: 2.147091325
   Max 1st sample XO time among all SL packets: 45.256996999
   TM values from -349 to 278 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-26T21:10:36.147.089.725 (XO: 2.147091325 - 2.147397326, t=45.2 degC)  Rank: 00073
   TM values: Max =   278  Min =  -274  Ave =    -4.41  Std =    43.60
   Min TM value:-274 at an offset of 128.225 usec -> 2023-06-26T21:10:36.147.217.950 (XO: 2.147219551)
   Max TM value:278 at an offset of 130.850 usec -> 2023-06-26T21:10:36.147.220.575 (XO: 2.147222176)
   Max dTM/dt [1/us]:2.54e+03 at an offset of 158.238 usec -> 2023-06-26T21:10:36.147.247.963 (XO: 2.147249563)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 100.375 usec -> 2023-06-26T21:10:36.147.190.100 (XO: 2.147191700)

 - Block number: 2 starting at 2023-06-26T21:10:54.916.876.725 (XO: 20.916882342 - 20.917188343, t=45.2 degC)  Rank: 00082
   TM values: Max =   262  Min =  -349  Ave =    -4.50  Std =    40.49
   Min TM value:-349 at an offset of 126.525 usec -> 2023-06-26T21:10:54.917.003.250 (XO: 20.917008868)
   Max TM value:262 at an offset of 181.551 usec -> 2023-06-26T21:10:54.917.058.275 (XO: 20.917063893)
   Max dTM/dt [1/us]:2.78e+03 at an offset of 100.000 usec -> 2023-06-26T21:10:54.916.976.725 (XO: 20.916982343)
   Min dTM/dt [1/us]:-3.17e+03 at an offset of 124.188 usec -> 2023-06-26T21:10:54.917.000.912 (XO: 20.917006530)

 - Block number: 3 starting at 2023-06-26T21:11:05.876.849.725 (XO: 31.876855106 - 31.877161107, t=45.2 degC)  Rank: 00079
   TM values: Max =   222  Min =  -326  Ave =    -4.49  Std =    27.44
   Min TM value:-326 at an offset of 103.963 usec -> 2023-06-26T21:11:05.876.953.687 (XO: 31.876959069)
   Max TM value:222 at an offset of 101.488 usec -> 2023-06-26T21:11:05.876.951.212 (XO: 31.876956594)
   Max dTM/dt [1/us]:2.37e+03 at an offset of 104.113 usec -> 2023-06-26T21:11:05.876.953.837 (XO: 31.876959219)
   Min dTM/dt [1/us]:-3.10e+03 at an offset of 101.625 usec -> 2023-06-26T21:11:05.876.951.350 (XO: 31.876956732)

 - Block number: 4 starting at 2023-06-26T21:11:19.256.705.725 (XO: 45.256707998 - 45.257013999, t=45.2 degC)  Rank: 00079
   TM values: Max =   251  Min =  -312  Ave =    -4.56  Std =    29.54
   Min TM value:-312 at an offset of 128.075 usec -> 2023-06-26T21:11:19.256.833.800 (XO: 45.256836073)
   Max TM value:251 at an offset of 127.813 usec -> 2023-06-26T21:11:19.256.833.537 (XO: 45.256835811)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 130.325 usec -> 2023-06-26T21:11:19.256.836.050 (XO: 45.256838323)
   Min dTM/dt [1/us]:-2.95e+03 at an offset of 127.913 usec -> 2023-06-26T21:11:19.256.833.637 (XO: 45.256835911)
