-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jul  5 01:11:10 2023 UTC
-- File 20230626_204959_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.858924 to 27.119268
-each separately from 6s to 27s and from 118979us to 859213us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.259766 in 1 packets: 18
MIU packet time from 2023-06-26T20:50:08.858.924.000 to 2023-06-26T20:50:29.119.268.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.259766 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.7090503 to 28.7089801
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999298 in 1 packets: 18
Temperature FPGA: 
      FD -> 58.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      FE -> 57.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230626_204959_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5679767 to 8.5682657
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.2596950  -- OK for SL --
   MIU time from 2023-06-26T20:50:08.858.924.000 to 2023-06-26T20:50:08.859.213.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.259766

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8279607 to 28.8282497
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-26T20:50:29.118.979.000 to 2023-06-26T20:50:29.119.268.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.567976700 corrected by 40bit 10MHz counter of 7.709050300 obtained at 1Hz counter of 6. Result: 6.858929234
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-26T20:50:08.858.783.725
   Max 1st sample MIU time among all SL packets: 2023-06-26T20:50:29.119.127.725
   Min 1st sample XO time among all SL packets: 6.858788959
   Max 1st sample XO time among all SL packets: 27.119129718
   TM values from -341 to 253 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-26T20:50:08.858.783.725 (XO: 6.858788959 - 6.859094960, t=45.2 degC)  Rank: 00081
   TM values: Max =   253  Min =  -315  Ave =    -4.37  Std =    28.40
   Min TM value:-315 at an offset of 119.825 usec -> 2023-06-26T20:50:08.858.903.550 (XO: 6.858908784)
   Max TM value:253 at an offset of 124.688 usec -> 2023-06-26T20:50:08.858.908.412 (XO: 6.858913647)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 122.238 usec -> 2023-06-26T20:50:08.858.905.962 (XO: 6.858911197)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 119.650 usec -> 2023-06-26T20:50:08.858.903.375 (XO: 6.858908609)

 - Block number: 2 starting at 2023-06-26T20:50:29.118.838.725 (XO: 27.118840717 - 27.119146718, t=45.2 degC)  Rank: 00084
   TM values: Max =   242  Min =  -341  Ave =    -4.41  Std =    38.86
   Min TM value:-341 at an offset of 121.113 usec -> 2023-06-26T20:50:29.118.959.837 (XO: 27.118961830)
   Max TM value:242 at an offset of 118.600 usec -> 2023-06-26T20:50:29.118.957.325 (XO: 27.118959318)
   Max dTM/dt [1/us]:2.55e+03 at an offset of 121.188 usec -> 2023-06-26T20:50:29.118.959.912 (XO: 27.118961905)
   Min dTM/dt [1/us]:-2.60e+03 at an offset of 118.800 usec -> 2023-06-26T20:50:29.118.957.525 (XO: 27.118959518)
