-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jul  5 00:12:09 2023 UTC
-- File 20230626_190617_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.798563 to 15.338922
-each separately from 4s to 15s and from 338633us to 798852us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.539781 in 1 packets: 18
MIU packet time from 2023-06-26T19:06:24.798.563.000 to 2023-06-26T19:06:35.338.922.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.539781 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.7236183 to 16.7235814
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999631 in 1 packets: 18
Temperature FPGA: 
      FE -> 57.6 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230626_190617_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5221837 to 6.5224727
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5397430  -- OK for SL --
   MIU time from 2023-06-26T19:06:24.798.563.000 to 2023-06-26T19:06:24.798.852.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.539781

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0622157 to 17.0625047
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-26T19:06:35.338.633.000 to 2023-06-26T19:06:35.338.922.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.522183700 corrected by 40bit 10MHz counter of 5.723618300 obtained at 1Hz counter of 4. Result: 4.798568115
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-26T19:06:24.798.422.725
   Max 1st sample MIU time among all SL packets: 2023-06-26T19:06:35.338.781.725
   Min 1st sample XO time among all SL packets: 4.798427840
   Max 1st sample XO time among all SL packets: 15.338784177
   TM values from -384 to 323 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-26T19:06:24.798.422.725 (XO: 4.798427840 - 4.798733841, t=45.2 degC)  Rank: 00068
   TM values: Max =   257  Min =  -216  Ave =    -4.35  Std =    31.98
   Min TM value:-216 at an offset of 102.663 usec -> 2023-06-26T19:06:24.798.525.387 (XO: 4.798530503)
   Max TM value:257 at an offset of 119.225 usec -> 2023-06-26T19:06:24.798.541.950 (XO: 4.798547065)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 116.825 usec -> 2023-06-26T19:06:24.798.539.550 (XO: 4.798544665)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 102.525 usec -> 2023-06-26T19:06:24.798.525.250 (XO: 4.798530365)

 - Block number: 2 starting at 2023-06-26T19:06:35.338.492.725 (XO: 15.338495176 - 15.338801177, t=45.2 degC)  Rank: 00084
   TM values: Max =   323  Min =  -384  Ave =    -4.47  Std =    42.91
   Min TM value:-384 at an offset of 130.800 usec -> 2023-06-26T19:06:35.338.623.525 (XO: 15.338625976)
   Max TM value:323 at an offset of 128.388 usec -> 2023-06-26T19:06:35.338.621.112 (XO: 15.338623564)
   Max dTM/dt [1/us]:2.69e+03 at an offset of 130.863 usec -> 2023-06-26T19:06:35.338.623.587 (XO: 15.338626039)
   Min dTM/dt [1/us]:-3.04e+03 at an offset of 128.488 usec -> 2023-06-26T19:06:35.338.621.212 (XO: 15.338623664)
