-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jul  4 23:53:05 2023 UTC
-- File 20230626_183011_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.904780 to 24.925327
-each separately from 9s to 24s and from 904780us to 925327us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.019969 in 1 packets: 18
MIU packet time from 2023-06-26T18:30:23.904.780.000 to 2023-06-26T18:30:38.925.327.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.019969 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7048350 to 25.7047849
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999499 in 1 packets: 18
Temperature FPGA: 
      FE -> 57.6 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230626_183011_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6096176 to 11.6099066
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0199190  -- OK for SL --
   MIU time from 2023-06-26T18:30:23.904.780.000 to 2023-06-26T18:30:23.905.069.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.019969

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6298256 to 26.6301146
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-26T18:30:38.925.038.000 to 2023-06-26T18:30:38.925.327.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.609617600 corrected by 40bit 10MHz counter of 10.704835000 obtained at 1Hz counter of 9. Result: 9.904785586
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-26T18:30:23.904.639.725
   Max 1st sample MIU time among all SL packets: 2023-06-26T18:30:38.925.186.725
   Min 1st sample XO time among all SL packets: 9.904645310
   Max 1st sample XO time among all SL packets: 24.925192571
   TM values from -372 to 281 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-26T18:30:23.904.639.725 (XO: 9.904645310 - 9.904951311, t=45.2 degC)  Rank: 00077
   TM values: Max =   281  Min =  -315  Ave =    -4.37  Std =    37.95
   Min TM value:-315 at an offset of 119.938 usec -> 2023-06-26T18:30:23.904.759.662 (XO: 9.904765248)
   Max TM value:281 at an offset of 169.263 usec -> 2023-06-26T18:30:23.904.808.988 (XO: 9.904814573)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 122.413 usec -> 2023-06-26T18:30:23.904.762.137 (XO: 9.904767723)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 119.825 usec -> 2023-06-26T18:30:23.904.759.550 (XO: 9.904765136)

 - Block number: 2 starting at 2023-06-26T18:30:38.924.897.725 (XO: 24.924903570 - 24.925209571, t=45.2 degC)  Rank: 00081
   TM values: Max =   261  Min =  -372  Ave =    -4.45  Std =    34.27
   Min TM value:-372 at an offset of 117.925 usec -> 2023-06-26T18:30:38.925.015.650 (XO: 24.925021495)
   Max TM value:261 at an offset of 104.875 usec -> 2023-06-26T18:30:38.925.002.600 (XO: 24.925008445)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 120.275 usec -> 2023-06-26T18:30:38.925.018.000 (XO: 24.925023845)
   Min dTM/dt [1/us]:-2.90e+03 at an offset of 117.775 usec -> 2023-06-26T18:30:38.925.015.500 (XO: 24.925021345)
