-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 05:09:57 2023 UTC
-- File 20230604_165217_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.778571 to 14.778860
-each separately from 14s to 14s and from 778571us to 778860us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2023-06-04T16:52:34.778.571.000 to 2023-06-04T16:52:34.778.860.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7628187 to 15.7628187
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_165217_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5413906 to 16.5416796
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T16:52:34.778.571.000 to 2023-06-04T16:52:34.778.860.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.541390600 corrected by 40bit 10MHz counter of 14.762822300 obtained at 1Hz counter of 13. Result: 14.778574703
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-04T16:52:34.778.430.725
   Max 1st sample MIU time among all SL packets: 2023-06-04T16:52:34.778.719.725
   Min 1st sample XO time among all SL packets: 14.778434427
   Max 1st sample XO time among all SL packets: 14.778723428
   TM values from -269 to 267 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-04T16:52:34.778.430.725 (XO: 14.778434427 - 14.778740428, t=40.4 degC)  Rank: 00073
   TM values: Max =   267  Min =  -269  Ave =    -4.42  Std =    60.78
   Min TM value:-269 at an offset of 127.175 usec -> 2023-06-04T16:52:34.778.557.900 (XO: 14.778561603)
   Max TM value:267 at an offset of 170.226 usec -> 2023-06-04T16:52:34.778.600.950 (XO: 14.778604653)
   Max dTM/dt [1/us]:2.75e+03 at an offset of 129.788 usec -> 2023-06-04T16:52:34.778.560.512 (XO: 14.778564215)
   Min dTM/dt [1/us]:-2.55e+03 at an offset of 99.763 usec -> 2023-06-04T16:52:34.778.530.487 (XO: 14.778534190)
