-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 03:14:44 2023 UTC
-- File 20230604_134430_1229.dat, 905  packets found.
APID:
      1229 (EW) in 905 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 905 packets (all packets)
Total number of packets: 
      36720 in 905 packets (all packets)
Packet No from 1 to 905
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 5.492460 to 5.492460
-each separately from 5s to 5s and from 492460us to 492460us.
   Differences in neigboring packets: 
      0.000000 in 904 packets (all packets)
MIU packet time from 2023-06-04T13:44:38.492.460.000 to 2023-06-04T13:44:38.492.460.000
   Differences in neigboring packets: 
      0.000000 in 904 packets (all packets)
MIU orbit number: 
      60000 in 905 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 905 packets (all packets)
MIU orbit information / number of events: 
      0 in 905 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 905 packets (all packets)
Instrument anomalies: 
      0 (OK) in 905 packets (all packets)
FPGA configuration version: 
      211 in 905 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9234957 to 6.9234957
   Differences in neigboring packets: 
      0.0000000 in 904 packets (all packets)
Temperature FPGA: 
      03 -> 51.1 degC in 905 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 905 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 905 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 905 packets (all packets)
Number of external events: 
      2 in 905 packets (all packets)
Number of generated alerts: 
      2 in 905 packets (all packets)
Survey memory error: 
      0 in 905 packets (all packets)
Event memory error: 
      0 in 905 packets (all packets)
Configuration error: 
      0 in 905 packets (all packets)
SDRAM CRC error: 
      0 in 905 packets (all packets)
Radio gain: 
      0 in 905 packets (all packets)
Radio bandwidth: 
      0 in 905 packets (all packets)
Digital gain: 
      0 in 905 packets (all packets)
Analog gain: 
      0 in 905 packets (all packets)

EVENT DATA READER
-- File 20230604_134430_1229.dat, 4 EW pages found. Extra packets: 89.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 2 to 2
      2 in 816 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 7.5026569 to 7.5165119
      Difference between the neighboring packets: 
      0.0000170 in 815 packets (all packets)
   Memory page number from 0 to 3
      Difference between the neighboring packets: 
      0 in 812 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 3 packets: 204, 408, 612
   Event page number from 42 to 45
      Difference between the neighboring packets: 
      0 in 812 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 3 packets: 204, 408, 612
   MIU event flag 
      1 in 816 packets (all packets)
   Time triggered event flag 
      0 in 816 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-06-04T13:44:38.436.971.825
   Min EW header abs. time found among all events            : 2023-06-04T13:44:38.492.460.000
   Max EW event (1st sample) abs. time found among all events: 2023-06-04T13:44:38.436.971.825
   Max EW header abs. time found among all events            : 2023-06-04T13:44:38.492.460.000
   TM values from -226 to 218 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 816
    Page no. 42:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 4  ordered packets: 816
   TM values from -226 to 218 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=5 C1MHz=0.492460
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 815 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 7.5026569
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-06-04T13:44:38.492.460.000
   Event 1st sample abs. time : 2023-06-04T13:44:38.436.971.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 from 2023-06-04T13:44:38.436.971.825 to 2023-06-04T13:44:38.450.843.825 (XO: -333.000000175 - -332.986128175)
   Min TM value:-226 at an offset of 125.412 usec -> 2023-06-04T13:44:38.437.097.237 (XO: -332.999874763)
   Max TM value:218 at an offset of 54.312 usec -> 2023-06-04T13:44:38.437.026.137 (XO: -332.999945862)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 127.875 usec -> 2023-06-04T13:44:38.437.099.700 (XO: -332.999872300)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 54.413 usec -> 2023-06-04T13:44:38.437.026.237 (XO: -332.999945763)
   Med: -5.00000   Ave: -4.5418550   Std: 5.5252982
