-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 02:41:25 2023 UTC
-- File 20230604_131637_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.053706 to 22.870206
-each separately from 8s to 22s and from 53706us to 870206us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.815922 in 1 packets: 18
MIU packet time from 2023-06-04T13:16:48.053.706.000 to 2023-06-04T13:17:02.870.206.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.815922 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8788744 to 23.8788240
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999496 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      02 -> 52.4 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 11 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30
      4 in 6 packets: 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_131637_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.9325820 to 9.9328710
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8158710  -- OK for SL --
   MIU time from 2023-06-04T13:16:48.053.706.000 to 2023-06-04T13:16:48.053.995.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.815922

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.7487420 to 24.7490310
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T13:17:02.869.917.000 to 2023-06-04T13:17:02.870.206.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.932582000 corrected by 40bit 10MHz counter of 9.878874400 obtained at 1Hz counter of 8. Result: 8.053707793
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-04T13:16:48.053.565.725
   Max 1st sample MIU time among all SL packets: 2023-06-04T13:17:02.870.065.725
   Min 1st sample XO time among all SL packets: 8.053567518
   Max 1st sample XO time among all SL packets: 22.870069857
   TM values from -863 to 371 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-04T13:16:48.053.565.725 (XO: 8.053567518 - 8.053873519, t=40.4 degC)  Rank: 00084
   TM values: Max =   216  Min =  -272  Ave =    -4.45  Std =    27.71
   Min TM value:-272 at an offset of 120.025 usec -> 2023-06-04T13:16:48.053.685.750 (XO: 8.053687543)
   Max TM value:216 at an offset of 119.750 usec -> 2023-06-04T13:16:48.053.685.475 (XO: 8.053687268)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 150.851 usec -> 2023-06-04T13:16:48.053.716.575 (XO: 8.053718368)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 119.900 usec -> 2023-06-04T13:16:48.053.685.625 (XO: 8.053687418)

 - Block number: 2 starting at 2023-06-04T13:17:02.869.776.725 (XO: 22.869780856 - 22.870086857, t=40.4 degC)  Rank: 00091
   TM values: Max =   371  Min =  -863  Ave =    -3.95  Std =    53.20
   Min TM value:-863 at an offset of 291.114 usec -> 2023-06-04T13:17:02.870.067.838 (XO: 22.870071970)
   Max TM value:371 at an offset of 296.164 usec -> 2023-06-04T13:17:02.870.072.888 (XO: 22.870077020)
   Max dTM/dt [1/us]:5.68e+03 at an offset of 70.713 usec -> 2023-06-04T13:17:02.869.847.437 (XO: 22.869851569)
   Min dTM/dt [1/us]:-5.82e+03 at an offset of 70.613 usec -> 2023-06-04T13:17:02.869.847.337 (XO: 22.869851469)
