-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 00:49:01 2023 UTC
-- File 20230604_101303_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.469223 to 18.249706
-each separately from 13s to 18s and from 249417us to 469512us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.779905 in 1 packets: 18
MIU packet time from 2023-06-04T10:13:19.469.223.000 to 2023-06-04T10:13:24.249.706.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.779905 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9310201 to 19.9310020
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999819 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_101303_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.4002439 to 15.4005329
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.7798870  -- OK for SL --
   MIU time from 2023-06-04T10:13:19.469.223.000 to 2023-06-04T10:13:19.469.512.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.779905

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.1804199 to 20.1807089
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T10:13:24.249.417.000 to 2023-06-04T10:13:24.249.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.400243900 corrected by 40bit 10MHz counter of 14.931020100 obtained at 1Hz counter of 13. Result: 13.469225536
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-04T10:13:19.469.082.725
   Max 1st sample MIU time among all SL packets: 2023-06-04T10:13:24.249.565.725
   Min 1st sample XO time among all SL packets: 13.469085261
   Max 1st sample XO time among all SL packets: 18.249567523
   TM values from -235 to 186 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-04T10:13:19.469.082.725 (XO: 13.469085261 - 13.469391262, t=40.4 degC)  Rank: 00084
   TM values: Max =   186  Min =  -235  Ave =    -4.45  Std =    19.17
   Min TM value:-235 at an offset of 122.763 usec -> 2023-06-04T10:13:19.469.205.487 (XO: 13.469208024)
   Max TM value:186 at an offset of 125.413 usec -> 2023-06-04T10:13:19.469.208.137 (XO: 13.469210674)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 125.250 usec -> 2023-06-04T10:13:19.469.207.975 (XO: 13.469210511)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 122.600 usec -> 2023-06-04T10:13:19.469.205.325 (XO: 13.469207861)

 - Block number: 2 starting at 2023-06-04T10:13:24.249.276.725 (XO: 18.249278522 - 18.249584524, t=40.4 degC)  Rank: 00087
   TM values: Max =   172  Min =  -214  Ave =    -4.46  Std =    17.95
   Min TM value:-214 at an offset of 125.675 usec -> 2023-06-04T10:13:24.249.402.400 (XO: 18.249404198)
   Max TM value:172 at an offset of 110.363 usec -> 2023-06-04T10:13:24.249.387.087 (XO: 18.249388885)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 125.800 usec -> 2023-06-04T10:13:24.249.402.525 (XO: 18.249404323)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 110.500 usec -> 2023-06-04T10:13:24.249.387.225 (XO: 18.249389023)
