-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 00:47:00 2023 UTC
-- File 20230604_101120_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.105865 to 28.076689
-each separately from 13s to 28s and from 76400us to 106154us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.970246 in 1 packets: 18
MIU packet time from 2023-06-04T10:11:36.105.865.000 to 2023-06-04T10:11:51.076.689.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.970246 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8309994 to 29.8309452
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999458 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_101120_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.9368660 to 14.9371550
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9701910  -- OK for SL --
   MIU time from 2023-06-04T10:11:36.105.865.000 to 2023-06-04T10:11:36.106.154.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.970246

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.9073460 to 29.9076350
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T10:11:51.076.400.000 to 2023-06-04T10:11:51.076.689.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.936866000 corrected by 40bit 10MHz counter of 14.830999400 obtained at 1Hz counter of 13. Result: 13.105866981
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-04T10:11:36.105.724.725
   Max 1st sample MIU time among all SL packets: 2023-06-04T10:11:51.076.548.725
   Min 1st sample XO time among all SL packets: 13.105726706
   Max 1st sample XO time among all SL packets: 28.076549801
   TM values from -234 to 184 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-04T10:11:36.105.724.725 (XO: 13.105726706 - 13.106032707, t=40.4 degC)  Rank: 00078
   TM values: Max =   184  Min =  -222  Ave =    -4.36  Std =    17.54
   Min TM value:-222 at an offset of 124.663 usec -> 2023-06-04T10:11:36.105.849.387 (XO: 13.105851369)
   Max TM value:184 at an offset of 127.375 usec -> 2023-06-04T10:11:36.105.852.100 (XO: 13.105854081)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 127.238 usec -> 2023-06-04T10:11:36.105.851.962 (XO: 13.105853944)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 124.500 usec -> 2023-06-04T10:11:36.105.849.225 (XO: 13.105851206)

 - Block number: 2 starting at 2023-06-04T10:11:51.076.259.725 (XO: 28.076260800 - 28.076566801, t=40.4 degC)  Rank: 00083
   TM values: Max =   178  Min =  -234  Ave =    -4.56  Std =    24.16
   Min TM value:-234 at an offset of 123.638 usec -> 2023-06-04T10:11:51.076.383.362 (XO: 28.076384437)
   Max TM value:178 at an offset of 78.850 usec -> 2023-06-04T10:11:51.076.338.575 (XO: 28.076339650)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 123.750 usec -> 2023-06-04T10:11:51.076.383.475 (XO: 28.076384550)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 121.075 usec -> 2023-06-04T10:11:51.076.380.800 (XO: 28.076381875)
