-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 00:36:25 2023 UTC
-- File 20230604_100219_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177029
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 6 packets: 2, 5, 8, 11, 15, 17
      0.833944 in 8 packets: 3, 4, 6, 9, 10, 12, 14, 16
      0.833940 in 2 packets: 7, 13
MIU packet time from 2023-06-04T10:02:22.000.000.000 to 2023-06-04T10:02:36.177.029.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 6 packets: 2, 5, 8, 11, 15, 17
      0.833944 in 8 packets: 3, 4, 6, 9, 10, 12, 14, 16
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8462196 to 15.8461692
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999964 in 10 packets: 2, 4, 6, 8, 9, 10, 11, 12, 14, 16
      0.9999965 in 2 packets: 3, 17
      0.9999963 in 2 packets: 5, 15
   XO PPM: 
      -3.6 in 10 packets: 2, 4, 6, 8, 9, 10, 11, 12, 14, 16
      -3.5 in 2 packets: 3, 17
      -3.7 in 2 packets: 5, 15
Temperature FPGA: 
      02 -> 52.4 degC in 15 packets: 1, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 17, 18
      01 -> 53.7 degC in 3 packets: 2, 15, 16
Temperature XO: 
      C3 -> 40.4 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 11 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11
      1 in 7 packets: 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 11 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11
      1 in 7 packets: 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_100219_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8462197 to 16.0231997
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T10:02:22.000.000.000 to 2023-06-04T10:02:36.177.029.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 6 packets: 2, 5, 8, 11, 15, 17
      0.833944 in 8 packets: 3, 4, 6, 9, 10, 12, 14, 16
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.846219700 corrected by 40bit 10MHz counter of 1.846219600 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996300 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2023-06-04T10:02:21.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-06-04T10:02:36.177.028.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031045
   TM values from -17 to 10 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-06-04T10:02:21.999.999.925 (XO: 0.000000025 - 0.000017025, t=40.4 degC)
   TM values: Max =     4  Min =   -14  Ave =    -4.73  Std =     3.08

 - Packet number: 2 starting at 2023-06-04T10:02:22.833.938.925 (XO: 0.833943027 - 0.833960027, t=40.4 degC)
   TM values: Max =     5  Min =   -16  Ave =    -4.52  Std =     3.23

 - Packet number: 3 starting at 2023-06-04T10:02:23.667.881.925 (XO: 1.667885963 - 1.667902963, t=40.4 degC)
   TM values: Max =     6  Min =   -15  Ave =    -4.52  Std =     3.50

 - Packet number: 4 starting at 2023-06-04T10:02:24.501.825.925 (XO: 2.501828932 - 2.501845932, t=40.4 degC)
   TM values: Max =     4  Min =   -15  Ave =    -4.55  Std =     3.05

 - Packet number: 5 starting at 2023-06-04T10:02:25.335.769.925 (XO: 3.335771967 - 3.335788967, t=40.4 degC)
   TM values: Max =    10  Min =   -14  Ave =    -4.40  Std =     3.23

 - Packet number: 6 starting at 2023-06-04T10:02:26.169.712.925 (XO: 4.169715036 - 4.169732036, t=40.4 degC)
   TM values: Max =     7  Min =   -13  Ave =    -4.20  Std =     2.91

 - Packet number: 7 starting at 2023-06-04T10:02:27.003.656.925 (XO: 5.003658038 - 5.003675038, t=40.4 degC)
   TM values: Max =     5  Min =   -17  Ave =    -4.49  Std =     3.33

 - Packet number: 8 starting at 2023-06-04T10:02:27.837.596.925 (XO: 5.837601040 - 5.837618040, t=40.4 degC)
   TM values: Max =     9  Min =   -14  Ave =    -4.62  Std =     3.31

 - Packet number: 9 starting at 2023-06-04T10:02:28.671.539.925 (XO: 6.671544043 - 6.671561043, t=40.4 degC)
   TM values: Max =     6  Min =   -14  Ave =    -4.46  Std =     2.98

 - Packet number: 10 starting at 2023-06-04T10:02:29.505.483.925 (XO: 7.505487045 - 7.505504045, t=40.4 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.51  Std =     2.60

 - Packet number: 11 starting at 2023-06-04T10:02:30.339.427.925 (XO: 8.339430047 - 8.339447047, t=40.4 degC)
   TM values: Max =     7  Min =   -14  Ave =    -4.45  Std =     2.99

 - Packet number: 12 starting at 2023-06-04T10:02:31.173.370.925 (XO: 9.173373049 - 9.173390049, t=40.4 degC)
   TM values: Max =     4  Min =   -14  Ave =    -4.33  Std =     2.83

 - Packet number: 13 starting at 2023-06-04T10:02:32.007.314.925 (XO: 10.007316051 - 10.007333051, t=40.4 degC)
   TM values: Max =     3  Min =   -13  Ave =    -4.59  Std =     2.55

 - Packet number: 14 starting at 2023-06-04T10:02:32.841.254.925 (XO: 10.841259054 - 10.841276054, t=40.4 degC)
   TM values: Max =     7  Min =   -15  Ave =    -4.56  Std =     3.27

 - Packet number: 15 starting at 2023-06-04T10:02:33.675.198.925 (XO: 11.675202123 - 11.675219123, t=40.4 degC)
   TM values: Max =     7  Min =   -13  Ave =    -4.22  Std =     2.57

 - Packet number: 16 starting at 2023-06-04T10:02:34.509.141.925 (XO: 12.509145158 - 12.509162158, t=40.4 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.63  Std =     3.47

 - Packet number: 17 starting at 2023-06-04T10:02:35.343.085.925 (XO: 13.343088126 - 13.343105126, t=40.4 degC)
   TM values: Max =     5  Min =   -15  Ave =    -4.20  Std =     3.14

 - Packet number: 18 starting at 2023-06-04T10:02:36.177.028.925 (XO: 14.177031045 - 14.177048045, t=40.4 degC)
   TM values: Max =     4  Min =   -15  Ave =    -4.58  Std =     2.82
