-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jun  4 22:38:05 2023 UTC
-- File 20230604_045343_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.363490 to 32.726169
-each separately from 8s to 32s and from 56179us to 726169us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.692400 in 1 packets: 18
      15.669412 in 1 packets: 36
MIU packet time from 2023-06-04T04:53:54.363.490.000 to 2023-06-04T04:54:18.726.169.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.692400 in 1 packets: 18
      15.669412 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.7395761 to 33.7394903
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999678 in 1 packets: 18
      14.9999464 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      4 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_045343_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.1030676 to 10.1033566
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.6923670  -- OK for SL --
   MIU time from 2023-06-04T04:53:54.363.490.000 to 2023-06-04T04:53:54.363.779.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.692400

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      234 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7957236 to 18.7960126
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6693590  -- OK for SL --
   MIU time from 2023-06-04T04:54:03.056.179.000 to 2023-06-04T04:54:03.056.468.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.669412

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      60 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4653716 to 34.4656606
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T04:54:18.725.880.000 to 2023-06-04T04:54:18.726.169.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.103067600 corrected by 40bit 10MHz counter of 9.739576100 obtained at 1Hz counter of 8. Result: 8.363492809
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-04T04:53:54.363.349.725
   Max 1st sample MIU time among all SL packets: 2023-06-04T04:54:18.726.028.725
   Min 1st sample XO time among all SL packets: 8.363352533
   Max 1st sample XO time among all SL packets: 32.726032639
   TM values from -425 to 1222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-04T04:53:54.363.349.725 (XO: 8.363352533 - 8.363658534, t=40.4 degC)  Rank: 00080
   TM values: Max =   442  Min =  -298  Ave =    -4.48  Std =    41.31
   Min TM value:-298 at an offset of 91.863 usec -> 2023-06-04T04:53:54.363.441.587 (XO: 8.363444396)
   Max TM value:442 at an offset of 91.338 usec -> 2023-06-04T04:53:54.363.441.062 (XO: 8.363443871)
   Max dTM/dt [1/us]:4.80e+03 at an offset of 91.300 usec -> 2023-06-04T04:53:54.363.441.025 (XO: 8.363443833)
   Min dTM/dt [1/us]:-4.76e+03 at an offset of 46.563 usec -> 2023-06-04T04:53:54.363.396.287 (XO: 8.363399096)

 - Block number: 2 starting at 2023-06-04T04:54:03.056.038.725 (XO: 17.056039627 - 17.056345628, t=40.4 degC)  Rank: 00234
   TM values: Max =  1222  Min =  -425  Ave =    -4.56  Std =    71.20
   Min TM value:-425 at an offset of 147.801 usec -> 2023-06-04T04:54:03.056.186.525 (XO: 17.056187427)
   Max TM value:1222 at an offset of 90.625 usec -> 2023-06-04T04:54:03.056.129.350 (XO: 17.056130252)
   Max dTM/dt [1/us]:7.22e+03 at an offset of 90.563 usec -> 2023-06-04T04:54:03.056.129.287 (XO: 17.056130190)
   Min dTM/dt [1/us]:-5.63e+03 at an offset of 130.175 usec -> 2023-06-04T04:54:03.056.168.900 (XO: 17.056169802)

 - Block number: 3 starting at 2023-06-04T04:54:18.725.739.725 (XO: 32.725743638 - 32.726049639, t=40.4 degC)  Rank: 00060
   TM values: Max =   228  Min =  -273  Ave =    -4.48  Std =    24.15
   Min TM value:-273 at an offset of 127.575 usec -> 2023-06-04T04:54:18.725.867.300 (XO: 32.725871213)
   Max TM value:228 at an offset of 127.300 usec -> 2023-06-04T04:54:18.725.867.025 (XO: 32.725870938)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 129.925 usec -> 2023-06-04T04:54:18.725.869.650 (XO: 32.725873563)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 127.425 usec -> 2023-06-04T04:54:18.725.867.150 (XO: 32.725871063)
