-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jun  4 07:40:05 2023 UTC
-- File 20230604_021609_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.307383 to 25.427744
-each separately from 0s to 25s and from 307383us to 427744us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.119783 in 1 packets: 18
MIU packet time from 2023-06-04T02:16:12.307.383.000 to 2023-06-04T02:16:37.427.744.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.119783 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7868543 to 26.7867660
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999117 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C4 -> 42.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230604_021609_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0942382 to 2.0945272
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.1196950  -- OK for SL --
   MIU time from 2023-06-04T02:16:12.307.383.000 to 2023-06-04T02:16:12.307.672.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.119783

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.2142222 to 27.2145112
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-04T02:16:37.427.455.000 to 2023-06-04T02:16:37.427.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.094238200 corrected by 40bit 10MHz counter of 1.786854300 obtained at 1Hz counter of 0. Result: 0.307384976
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-04T02:16:12.307.242.725
   Max 1st sample MIU time among all SL packets: 2023-06-04T02:16:37.427.603.725
   Min 1st sample XO time among all SL packets: 0.307244700
   Max 1st sample XO time among all SL packets: 25.427606464
   TM values from -261 to 193 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-04T02:16:12.307.242.725 (XO: 0.307244700 - 0.307550701, t=40.4 degC)  Rank: 00087
   TM values: Max =   193  Min =  -261  Ave =    -4.45  Std =    19.83
   Min TM value:-261 at an offset of 109.813 usec -> 2023-06-04T02:16:12.307.352.537 (XO: 0.307354513)
   Max TM value:193 at an offset of 114.738 usec -> 2023-06-04T02:16:12.307.357.462 (XO: 0.307359438)
   Max dTM/dt [1/us]:1.37e+03 at an offset of 95.113 usec -> 2023-06-04T02:16:12.307.337.837 (XO: 0.307339813)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 97.413 usec -> 2023-06-04T02:16:12.307.340.137 (XO: 0.307342113)

 - Block number: 2 starting at 2023-06-04T02:16:37.427.314.725 (XO: 25.427317463 - 25.427623464, t=42.0 degC)  Rank: 00079
   TM values: Max =   170  Min =  -224  Ave =    -4.49  Std =    23.58
   Min TM value:-224 at an offset of 123.713 usec -> 2023-06-04T02:16:37.427.438.437 (XO: 25.427441176)
   Max TM value:170 at an offset of 163.963 usec -> 2023-06-04T02:16:37.427.478.688 (XO: 25.427481426)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 163.776 usec -> 2023-06-04T02:16:37.427.478.500 (XO: 25.427481239)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 123.563 usec -> 2023-06-04T02:16:37.427.438.287 (XO: 25.427441026)
