-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar 21 17:51:18 2023 UTC
-- File 20221016_084550_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.362386 to 48.080476
-each separately from 1s to 48s and from 80187us to 751174us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.038763 in 1 packets: 18
      11.349158 in 1 packets: 36
      14.329013 in 1 packets: 54
MIU packet time from 2022-10-16T08:45:54.362.386.000 to 2022-10-16T08:46:41.080.476.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.038763 in 1 packets: 18
      11.349158 in 1 packets: 36
      14.329013 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9590262 to 49.9588573
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999244 in 1 packets: 18
      10.9999606 in 1 packets: 36
      14.9999461 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20221016_084550_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3214132 to 3.3217022
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0386870  -- OK for SL --
   MIU time from 2022-10-16T08:45:54.362.386.000 to 2022-10-16T08:45:54.362.675.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.038763

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3603892 to 24.3606782
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.3491190  -- OK for SL --
   MIU time from 2022-10-16T08:46:15.401.438.000 to 2022-10-16T08:46:15.401.727.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.349158

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.7097972 to 35.7100862
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3289590  -- OK for SL --
   MIU time from 2022-10-16T08:46:26.750.885.000 to 2022-10-16T08:46:26.751.174.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.329013

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.0390452 to 50.0393342
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-10-16T08:46:41.080.187.000 to 2022-10-16T08:46:41.080.476.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.321413200 corrected by 40bit 10MHz counter of 2.959026200 obtained at 1Hz counter of 1. Result: 1.362388305
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-10-16T08:45:54.362.245.725
   Max 1st sample MIU time among all SL packets: 2022-10-16T08:46:41.080.335.725
   Min 1st sample XO time among all SL packets: 1.362248029
   Max 1st sample XO time among all SL packets: 48.080336914
   TM values from -248 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-10-16T08:45:54.362.245.725 (XO: 1.362248029 - 1.362554030, t=37.2 degC)  Rank: 00082
   TM values: Max =   201  Min =  -210  Ave =    -4.32  Std =    20.82
   Min TM value:-210 at an offset of 116.975 usec -> 2022-10-16T08:45:54.362.362.700 (XO: 1.362365005)
   Max TM value:201 at an offset of 116.688 usec -> 2022-10-16T08:45:54.362.362.412 (XO: 1.362364717)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 114.650 usec -> 2022-10-16T08:45:54.362.360.375 (XO: 1.362362680)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 116.813 usec -> 2022-10-16T08:45:54.362.362.537 (XO: 1.362364842)

 - Block number: 2 starting at 2022-10-16T08:46:15.401.297.725 (XO: 22.401299730 - 22.401605731, t=37.2 degC)  Rank: 00074
   TM values: Max =   167  Min =  -199  Ave =    -4.34  Std =    22.28
   Min TM value:-199 at an offset of 129.338 usec -> 2022-10-16T08:46:15.401.427.062 (XO: 22.401429068)
   Max TM value:167 at an offset of 116.975 usec -> 2022-10-16T08:46:15.401.414.700 (XO: 22.401416705)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 131.888 usec -> 2022-10-16T08:46:15.401.429.612 (XO: 22.401431618)
   Min dTM/dt [1/us]:-1.51e+03 at an offset of 129.150 usec -> 2022-10-16T08:46:15.401.426.875 (XO: 22.401428880)

 - Block number: 3 starting at 2022-10-16T08:46:26.750.744.725 (XO: 33.750748428 - 33.751054429, t=37.2 degC)  Rank: 00076
   TM values: Max =   198  Min =  -248  Ave =    -4.43  Std =    29.29
   Min TM value:-248 at an offset of 125.938 usec -> 2022-10-16T08:46:26.750.870.662 (XO: 33.750874366)
   Max TM value:198 at an offset of 123.913 usec -> 2022-10-16T08:46:26.750.868.637 (XO: 33.750872341)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 123.738 usec -> 2022-10-16T08:46:26.750.868.462 (XO: 33.750872166)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 125.825 usec -> 2022-10-16T08:46:26.750.870.550 (XO: 33.750874253)

 - Block number: 4 starting at 2022-10-16T08:46:41.080.046.725 (XO: 48.080047913 - 48.080353914, t=37.2 degC)  Rank: 00076
   TM values: Max =   182  Min =  -214  Ave =    -4.41  Std =    19.12
   Min TM value:-214 at an offset of 128.875 usec -> 2022-10-16T08:46:41.080.175.600 (XO: 48.080176789)
   Max TM value:182 at an offset of 116.813 usec -> 2022-10-16T08:46:41.080.163.537 (XO: 48.080164726)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 131.388 usec -> 2022-10-16T08:46:41.080.178.112 (XO: 48.080179301)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 128.725 usec -> 2022-10-16T08:46:41.080.175.450 (XO: 48.080176639)
