-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 05:15:20 2023 UTC
-- File 20220706_222843_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177030
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 6, 8, 10, 11, 14, 15, 17
      0.833943 in 5 packets: 3, 5, 9, 12, 16
      0.833940 in 2 packets: 7, 13
MIU packet time from 2022-07-06T22:28:46.000.000.000 to 2022-07-06T22:29:00.177.030.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 6, 8, 10, 11, 14, 15, 17
      0.833943 in 5 packets: 3, 5, 9, 12, 16
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2601620 to 16.2601114
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999964 in 12 packets: 2, 3, 4, 5, 8, 9, 10, 11, 12, 14, 15, 16
      0.9999963 in 2 packets: 6, 17
   XO PPM: 
      -3.6 in 12 packets: 2, 3, 4, 5, 8, 9, 10, 11, 12, 14, 15, 16
      -3.7 in 2 packets: 6, 17
Temperature FPGA: 
      03 -> 51.1 degC in 5 packets: 1, 9, 10, 12, 16
      04 -> 49.8 degC in 12 packets: 2, 3, 4, 5, 6, 7, 8, 11, 13, 14, 15, 18
      05 -> 48.5 degC in 1 packets: 17
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      1 in 2 packets: 17, 18
Number of generated alerts: 
      0 in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      1 in 2 packets: 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220706_222843_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2601621 to 16.4371421
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-06T22:28:46.000.000.000 to 2022-07-06T22:29:00.177.030.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 9 packets: 2, 4, 6, 8, 10, 11, 14, 15, 17
      0.833943 in 5 packets: 3, 5, 9, 12, 16
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.260162100 corrected by 40bit 10MHz counter of 2.260162000 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2022-07-06T22:28:45.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-07-06T22:29:00.177.029.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031280
   TM values from -46 to 39 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-07-06T22:28:45.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    36  Min =   -39  Ave =    -4.34  Std =    11.79

 - Packet number: 2 starting at 2022-07-06T22:28:46.833.938.925 (XO: 0.833943027 - 0.833960027, t=38.8 degC)
   TM values: Max =    27  Min =   -38  Ave =    -4.22  Std =    11.91

 - Packet number: 3 starting at 2022-07-06T22:28:47.667.882.925 (XO: 1.667886029 - 1.667903029, t=38.8 degC)
   TM values: Max =    32  Min =   -37  Ave =    -4.40  Std =    11.73

 - Packet number: 4 starting at 2022-07-06T22:28:48.501.825.925 (XO: 2.501829032 - 2.501846032, t=38.8 degC)
   TM values: Max =    27  Min =   -40  Ave =    -4.56  Std =    11.31

 - Packet number: 5 starting at 2022-07-06T22:28:49.335.769.925 (XO: 3.335772034 - 3.335789034, t=38.8 degC)
   TM values: Max =    32  Min =   -45  Ave =    -4.19  Std =    12.84

 - Packet number: 6 starting at 2022-07-06T22:28:50.169.712.925 (XO: 4.169715053 - 4.169732053, t=38.8 degC)
   TM values: Max =    38  Min =   -46  Ave =    -4.39  Std =    13.35

 - Packet number: 7 starting at 2022-07-06T22:28:51.003.656.925 (XO: 5.003658138 - 5.003675138, t=38.8 degC)
   TM values: Max =    34  Min =   -36  Ave =    -4.51  Std =    10.28

 - Packet number: 8 starting at 2022-07-06T22:28:51.837.596.925 (XO: 5.837601140 - 5.837618140, t=38.8 degC)
   TM values: Max =    32  Min =   -41  Ave =    -4.77  Std =    12.94

 - Packet number: 9 starting at 2022-07-06T22:28:52.671.540.925 (XO: 6.671544143 - 6.671561143, t=38.8 degC)
   TM values: Max =    33  Min =   -42  Ave =    -4.35  Std =    10.40

 - Packet number: 10 starting at 2022-07-06T22:28:53.505.483.925 (XO: 7.505487145 - 7.505504145, t=38.8 degC)
   TM values: Max =    32  Min =   -40  Ave =    -4.02  Std =    12.05

 - Packet number: 11 starting at 2022-07-06T22:28:54.339.427.925 (XO: 8.339430147 - 8.339447147, t=38.8 degC)
   TM values: Max =    39  Min =   -37  Ave =    -4.53  Std =    10.33

 - Packet number: 12 starting at 2022-07-06T22:28:55.173.371.925 (XO: 9.173373149 - 9.173390149, t=38.8 degC)
   TM values: Max =    26  Min =   -34  Ave =    -4.58  Std =     8.47

 - Packet number: 13 starting at 2022-07-06T22:28:56.007.314.925 (XO: 10.007316151 - 10.007333151, t=38.8 degC)
   TM values: Max =    27  Min =   -34  Ave =    -4.46  Std =     9.48

 - Packet number: 14 starting at 2022-07-06T22:28:56.841.254.925 (XO: 10.841259154 - 10.841276154, t=38.8 degC)
   TM values: Max =    25  Min =   -36  Ave =    -4.38  Std =     9.63

 - Packet number: 15 starting at 2022-07-06T22:28:57.675.198.925 (XO: 11.675202156 - 11.675219156, t=38.8 degC)
   TM values: Max =    28  Min =   -35  Ave =    -4.75  Std =    10.47

 - Packet number: 16 starting at 2022-07-06T22:28:58.509.142.925 (XO: 12.509145158 - 12.509162158, t=38.8 degC)
   TM values: Max =    20  Min =   -37  Ave =    -4.31  Std =     8.78

 - Packet number: 17 starting at 2022-07-06T22:28:59.343.085.925 (XO: 13.343088194 - 13.343105194, t=38.8 degC)
   TM values: Max =    29  Min =   -35  Ave =    -4.64  Std =     9.79

 - Packet number: 18 starting at 2022-07-06T22:29:00.177.029.925 (XO: 14.177031280 - 14.177048280, t=38.8 degC)
   TM values: Max =    33  Min =   -40  Ave =    -4.14  Std =    11.16
