-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 21:21:05 2023 UTC
-- File 20220703_214519_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.323217 to 17.734621
-each separately from 5s to 17s and from 323217us to 734621us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.410826 in 1 packets: 18
MIU packet time from 2022-07-03T21:45:27.323.217.000 to 2022-07-03T21:45:39.734.621.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.410826 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.1520367 to 19.1519939
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999572 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220703_214519_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4752547 to 7.4755437
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4107830  -- OK for SL --
   MIU time from 2022-07-03T21:45:27.323.217.000 to 2022-07-03T21:45:27.323.506.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.410826

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.8863267 to 19.8866157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-03T21:45:39.734.332.000 to 2022-07-03T21:45:39.734.621.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.475254700 corrected by 40bit 10MHz counter of 7.152036700 obtained at 1Hz counter of 5. Result: 5.323219131
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-07-03T21:45:27.323.076.725
   Max 1st sample MIU time among all SL packets: 2022-07-03T21:45:39.734.480.725
   Min 1st sample XO time among all SL packets: 5.323078856
   Max 1st sample XO time among all SL packets: 17.734484169
   TM values from -315 to 276 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-03T21:45:27.323.076.725 (XO: 5.323078856 - 5.323384857, t=38.8 degC)  Rank: 00084
   TM values: Max =   276  Min =  -315  Ave =    -4.44  Std =    33.22
   Min TM value:-315 at an offset of 125.563 usec -> 2022-07-03T21:45:27.323.202.287 (XO: 5.323204419)
   Max TM value:276 at an offset of 123.138 usec -> 2022-07-03T21:45:27.323.199.862 (XO: 5.323201994)
   Max dTM/dt [1/us]:2.85e+03 at an offset of 125.675 usec -> 2022-07-03T21:45:27.323.202.400 (XO: 5.323204531)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 123.263 usec -> 2022-07-03T21:45:27.323.199.987 (XO: 5.323202119)

 - Block number: 2 starting at 2022-07-03T21:45:39.734.191.725 (XO: 17.734195168 - 17.734501169, t=38.8 degC)  Rank: 00076
   TM values: Max =   263  Min =  -297  Ave =    -4.37  Std =    38.42
   Min TM value:-297 at an offset of 123.200 usec -> 2022-07-03T21:45:39.734.314.925 (XO: 17.734318369)
   Max TM value:263 at an offset of 137.750 usec -> 2022-07-03T21:45:39.734.329.475 (XO: 17.734332919)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 135.738 usec -> 2022-07-03T21:45:39.734.327.462 (XO: 17.734330906)
   Min dTM/dt [1/us]:-2.76e+03 at an offset of 137.875 usec -> 2022-07-03T21:45:39.734.329.600 (XO: 17.734333044)
