-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 21:18:27 2023 UTC
-- File 20220703_214357_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177029
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833943 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2022-07-03T21:44:00.000.000.000 to 2022-07-03T21:44:14.177.029.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833943 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2527059 to 16.2526559
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 4 packets: 2, 6, 12, 17
      0.9999964 in 10 packets: 3, 4, 5, 8, 9, 10, 11, 14, 15, 16
   XO PPM: 
      -3.5 in 4 packets: 2, 6, 12, 17
      -3.6 in 10 packets: 3, 4, 5, 8, 9, 10, 11, 14, 15, 16
Temperature FPGA: 
      02 -> 52.4 degC in 2 packets: 1, 5
      03 -> 51.1 degC in 13 packets: 2, 3, 4, 6, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 3 packets: 7, 8, 9
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 11 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17
      2 in 1 packets: 18
Number of generated alerts: 
      0 in 6 packets: 1, 2, 3, 4, 5, 6
      1 in 11 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17
      2 in 1 packets: 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220703_214357_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2527060 to 16.4296860
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-03T21:44:00.000.000.000 to 2022-07-03T21:44:14.177.029.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833943 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.252706000 corrected by 40bit 10MHz counter of 2.252705900 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2022-07-03T21:43:59.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-07-03T21:44:14.177.028.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177030645
   TM values from -80 to 68 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-07-03T21:43:59.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    49  Min =   -59  Ave =    -4.60  Std =    14.34

 - Packet number: 2 starting at 2022-07-03T21:44:00.833.938.925 (XO: 0.833942944 - 0.833959944, t=38.8 degC)
   TM values: Max =    34  Min =   -40  Ave =    -4.19  Std =    13.05

 - Packet number: 3 starting at 2022-07-03T21:44:01.667.882.925 (XO: 1.667885929 - 1.667902929, t=38.8 degC)
   TM values: Max =    52  Min =   -58  Ave =    -4.60  Std =    16.80

 - Packet number: 4 starting at 2022-07-03T21:44:02.501.825.925 (XO: 2.501828932 - 2.501845932, t=38.8 degC)
   TM values: Max =    45  Min =   -58  Ave =    -4.56  Std =    16.52

 - Packet number: 5 starting at 2022-07-03T21:44:03.335.769.925 (XO: 3.335771934 - 3.335788934, t=38.8 degC)
   TM values: Max =    55  Min =   -57  Ave =    -4.55  Std =    18.87

 - Packet number: 6 starting at 2022-07-03T21:44:04.169.712.925 (XO: 4.169714919 - 4.169731919, t=38.8 degC)
   TM values: Max =    44  Min =   -45  Ave =    -4.61  Std =    17.26

 - Packet number: 7 starting at 2022-07-03T21:44:05.003.656.925 (XO: 5.003657838 - 5.003674838, t=38.8 degC)
   TM values: Max =    68  Min =   -80  Ave =    -4.40  Std =    26.64

 - Packet number: 8 starting at 2022-07-03T21:44:05.837.596.925 (XO: 5.837600840 - 5.837617840, t=38.8 degC)
   TM values: Max =    61  Min =   -72  Ave =    -4.45  Std =    24.21

 - Packet number: 9 starting at 2022-07-03T21:44:06.671.540.925 (XO: 6.671543843 - 6.671560843, t=38.8 degC)
   TM values: Max =    56  Min =   -69  Ave =    -4.54  Std =    22.79

 - Packet number: 10 starting at 2022-07-03T21:44:07.505.483.925 (XO: 7.505486845 - 7.505503845, t=38.8 degC)
   TM values: Max =    53  Min =   -70  Ave =    -4.65  Std =    19.82

 - Packet number: 11 starting at 2022-07-03T21:44:08.339.427.925 (XO: 8.339429847 - 8.339446847, t=38.8 degC)
   TM values: Max =    53  Min =   -69  Ave =    -4.28  Std =    18.51

 - Packet number: 12 starting at 2022-07-03T21:44:09.173.370.925 (XO: 9.173372832 - 9.173389832, t=38.8 degC)
   TM values: Max =    43  Min =   -51  Ave =    -4.63  Std =    16.10

 - Packet number: 13 starting at 2022-07-03T21:44:10.007.314.925 (XO: 10.007315751 - 10.007332751, t=38.8 degC)
   TM values: Max =    48  Min =   -60  Ave =    -4.29  Std =    16.45

 - Packet number: 14 starting at 2022-07-03T21:44:10.841.254.925 (XO: 10.841258754 - 10.841275754, t=38.8 degC)
   TM values: Max =    57  Min =   -59  Ave =    -4.65  Std =    23.52

 - Packet number: 15 starting at 2022-07-03T21:44:11.675.197.925 (XO: 11.675201756 - 11.675218756, t=38.8 degC)
   TM values: Max =    36  Min =   -60  Ave =    -4.86  Std =    15.99

 - Packet number: 16 starting at 2022-07-03T21:44:12.509.141.925 (XO: 12.509144758 - 12.509161758, t=38.8 degC)
   TM values: Max =    40  Min =   -48  Ave =    -4.54  Std =    13.51

 - Packet number: 17 starting at 2022-07-03T21:44:13.343.085.925 (XO: 13.343087726 - 13.343104726, t=38.8 degC)
   TM values: Max =    42  Min =   -44  Ave =    -4.22  Std =    14.75

 - Packet number: 18 starting at 2022-07-03T21:44:14.177.028.925 (XO: 14.177030645 - 14.177047645, t=38.8 degC)
   TM values: Max =    49  Min =   -53  Ave =    -4.28  Std =    16.22
