-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 21:00:21 2023 UTC
-- File 20220703_211558_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.983429 to 15.342962
-each separately from 7s to 15s and from 342673us to 983718us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.358955 in 1 packets: 18
MIU packet time from 2022-07-03T21:16:08.983.429.000 to 2022-07-03T21:16:16.342.962.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.358955 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7593557 to 16.7593270
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999713 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220703_211558_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7427855 to 9.7430745
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.3589270  -- OK for SL --
   MIU time from 2022-07-03T21:16:08.983.429.000 to 2022-07-03T21:16:08.983.718.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.358955

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1020015 to 17.1022905
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-03T21:16:16.342.673.000 to 2022-07-03T21:16:16.342.962.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.742785500 corrected by 40bit 10MHz counter of 8.759355700 obtained at 1Hz counter of 7. Result: 7.983433340
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-07-03T21:16:08.983.288.725
   Max 1st sample MIU time among all SL packets: 2022-07-03T21:16:16.342.821.725
   Min 1st sample XO time among all SL packets: 7.983293065
   Max 1st sample XO time among all SL packets: 15.342824425
   TM values from -355 to 281 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-03T21:16:08.983.288.725 (XO: 7.983293065 - 7.983599066, t=38.8 degC)  Rank: 00082
   TM values: Max =   199  Min =  -248  Ave =    -4.53  Std =    27.01
   Min TM value:-248 at an offset of 121.638 usec -> 2022-07-03T21:16:08.983.410.362 (XO: 7.983414703)
   Max TM value:199 at an offset of 126.475 usec -> 2022-07-03T21:16:08.983.415.200 (XO: 7.983419540)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 124.025 usec -> 2022-07-03T21:16:08.983.412.750 (XO: 7.983417090)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 126.600 usec -> 2022-07-03T21:16:08.983.415.325 (XO: 7.983419665)

 - Block number: 2 starting at 2022-07-03T21:16:16.342.532.725 (XO: 15.342535424 - 15.342841425, t=38.8 degC)  Rank: 00082
   TM values: Max =   281  Min =  -355  Ave =    -4.56  Std =    37.90
   Min TM value:-355 at an offset of 115.313 usec -> 2022-07-03T21:16:16.342.648.037 (XO: 15.342650737)
   Max TM value:281 at an offset of 112.875 usec -> 2022-07-03T21:16:16.342.645.600 (XO: 15.342648299)
   Max dTM/dt [1/us]:2.62e+03 at an offset of 115.450 usec -> 2022-07-03T21:16:16.342.648.175 (XO: 15.342650874)
   Min dTM/dt [1/us]:-2.85e+03 at an offset of 112.925 usec -> 2022-07-03T21:16:16.342.645.650 (XO: 15.342648349)
