-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 19:55:57 2023 UTC
-- File 20220703_192204_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.815069 to 70.008775
-each separately from 9s to 70s and from 8486us to 954398us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.138751 in 1 packets: 18
      13.878337 in 1 packets: 36
      19.077618 in 1 packets: 54
      18.097555 in 1 packets: 72
MIU packet time from 2022-07-03T19:22:16.815.069.000 to 2022-07-03T19:23:17.008.775.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.138751 in 1 packets: 18
      13.878337 in 1 packets: 36
      19.077618 in 1 packets: 54
      18.097555 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.1075262 to 72.1073087
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999679 in 1 packets: 18
      13.9999501 in 1 packets: 36
      18.9999322 in 1 packets: 54
      18.9999323 in 1 packets: 72
Temperature FPGA: 
      03 -> 51.1 degC in 90 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220703_192204_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.9225956 to 11.9228846
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1387190  -- OK for SL --
   MIU time from 2022-07-03T19:22:16.815.069.000 to 2022-07-03T19:22:16.815.358.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.138751

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0616036 to 21.0618926
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8782870  -- OK for SL --
   MIU time from 2022-07-03T19:22:25.954.109.000 to 2022-07-03T19:22:25.954.398.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.878337

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9401796 to 34.9404686
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0775510  -- OK for SL --
   MIU time from 2022-07-03T19:22:39.832.735.000 to 2022-07-03T19:22:39.833.024.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.077618

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.0180196 to 54.0183086
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.0974870  -- OK for SL --
   MIU time from 2022-07-03T19:22:58.910.642.000 to 2022-07-03T19:22:58.910.931.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.097555

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 72.1157956 to 72.1160846
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-07-03T19:23:17.008.486.000 to 2022-07-03T19:23:17.008.775.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.922595600 corrected by 40bit 10MHz counter of 11.107526200 obtained at 1Hz counter of 9. Result: 9.815072334
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-07-03T19:22:16.814.928.725
   Max 1st sample MIU time among all SL packets: 2022-07-03T19:23:17.008.634.725
   Min 1st sample XO time among all SL packets: 9.814932059
   Max 1st sample XO time among all SL packets: 70.008635656
   TM values from -342 to 266 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-07-03T19:22:16.814.928.725 (XO: 9.814932059 - 9.815238060, t=38.8 degC)  Rank: 00086
   TM values: Max =   266  Min =  -323  Ave =    -4.44  Std =    32.52
   Min TM value:-323 at an offset of 125.975 usec -> 2022-07-03T19:22:16.815.054.700 (XO: 9.815058034)
   Max TM value:266 at an offset of 123.500 usec -> 2022-07-03T19:22:16.815.052.225 (XO: 9.815055559)
   Max dTM/dt [1/us]:2.74e+03 at an offset of 126.075 usec -> 2022-07-03T19:22:16.815.054.800 (XO: 9.815058134)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 123.600 usec -> 2022-07-03T19:22:16.815.052.325 (XO: 9.815055659)

 - Block number: 2 starting at 2022-07-03T19:22:25.953.968.725 (XO: 18.953972564 - 18.954278565, t=38.8 degC)  Rank: 00083
   TM values: Max =   257  Min =  -310  Ave =    -4.37  Std =    43.41
   Min TM value:-310 at an offset of 125.638 usec -> 2022-07-03T19:22:25.954.094.362 (XO: 18.954098202)
   Max TM value:257 at an offset of 159.001 usec -> 2022-07-03T19:22:25.954.127.725 (XO: 18.954131564)
   Max dTM/dt [1/us]:2.60e+03 at an offset of 125.750 usec -> 2022-07-03T19:22:25.954.094.475 (XO: 18.954098314)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 123.263 usec -> 2022-07-03T19:22:25.954.091.987 (XO: 18.954095827)

 - Block number: 3 starting at 2022-07-03T19:22:39.832.594.725 (XO: 32.832598122 - 32.832904123, t=38.8 degC)  Rank: 00078
   TM values: Max =   263  Min =  -291  Ave =    -4.44  Std =    35.43
   Min TM value:-291 at an offset of 86.800 usec -> 2022-07-03T19:22:39.832.681.525 (XO: 32.832684923)
   Max TM value:263 at an offset of 89.213 usec -> 2022-07-03T19:22:39.832.683.937 (XO: 32.832687335)
   Max dTM/dt [1/us]:2.64e+03 at an offset of 89.088 usec -> 2022-07-03T19:22:39.832.683.812 (XO: 32.832687210)
   Min dTM/dt [1/us]:-2.97e+03 at an offset of 86.675 usec -> 2022-07-03T19:22:39.832.681.400 (XO: 32.832684798)

 - Block number: 4 starting at 2022-07-03T19:22:58.910.501.725 (XO: 51.910506112 - 51.910812113, t=38.8 degC)  Rank: 00078
   TM values: Max =   257  Min =  -331  Ave =    -4.53  Std =    33.56
   Min TM value:-331 at an offset of 113.088 usec -> 2022-07-03T19:22:58.910.614.812 (XO: 51.910619200)
   Max TM value:257 at an offset of 69.413 usec -> 2022-07-03T19:22:58.910.571.137 (XO: 51.910575525)
   Max dTM/dt [1/us]:2.82e+03 at an offset of 113.175 usec -> 2022-07-03T19:22:58.910.614.900 (XO: 51.910619287)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 110.750 usec -> 2022-07-03T19:22:58.910.612.475 (XO: 51.910616862)

 - Block number: 5 starting at 2022-07-03T19:23:17.008.345.725 (XO: 70.008346655 - 70.008652656, t=38.8 degC)  Rank: 00077
   TM values: Max =   265  Min =  -342  Ave =    -4.46  Std =    39.13
   Min TM value:-342 at an offset of 128.413 usec -> 2022-07-03T19:23:17.008.474.137 (XO: 70.008475068)
   Max TM value:265 at an offset of 156.176 usec -> 2022-07-03T19:23:17.008.501.900 (XO: 70.008502831)
   Max dTM/dt [1/us]:2.73e+03 at an offset of 128.513 usec -> 2022-07-03T19:23:17.008.474.237 (XO: 70.008475168)
   Min dTM/dt [1/us]:-3.01e+03 at an offset of 126.100 usec -> 2022-07-03T19:23:17.008.471.825 (XO: 70.008472756)
