-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 13:47:53 2023 UTC
-- File 20220703_081058_1229.dat, 31008  packets found.
APID:
      1229 (EW) in 31008 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 31008 packets (all packets)
Total number of packets: 
      31008 in 31008 packets (all packets)
Packet No from 1 to 31008
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.296923 to 31.686411
-each separately from 9s to 31s and from 296923us to 686411us.
   Differences in neigboring packets: 
      0.000000 in 31003 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      22.389488 in 1 packets: 11832
      -0.110976 in 1 packets: 21624
      0.110976 in 1 packets: 28152
      -22.389488 in 1 packets: 30600
MIU packet time from 2022-07-03T08:11:10.296.923.000 to 2022-07-03T08:11:32.686.411.000
   Differences in neigboring packets: 
      0.000000 in 31003 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      22.389488 in 1 packets: 11832
      -0.110976 in 1 packets: 21624
      0.110976 in 1 packets: 28152
      -22.389488 in 1 packets: 30600
MIU orbit number: 
      60000 in 31008 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 31008 packets (all packets)
MIU orbit information / number of events: 
      0 in 31008 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 31008 packets (all packets)
Instrument anomalies: 
      0 (OK) in 31008 packets (all packets)
FPGA configuration version: 
      211 in 31008 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7641886 to 32.7641086
   Differences in neigboring packets: 
      0.0000000 in 31005 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      21.9999200 in 1 packets: 11832
      -21.9999200 in 1 packets: 30600
Temperature FPGA: 
      04 -> 49.8 degC in 31008 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 31008 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 31008 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 31008 packets (all packets)
Number of external events: 
      1 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 12240 packets: 11833, 11834, 11835, 11836, 11837, 11838, 11839, 11840, 11841, 11842, 11843, 11844, 11845, 11846, 11847, 11848, 11849, 11850, ...
      2 in 6528 packets: 21625, 21626, 21627, 21628, 21629, 21630, 21631, 21632, 21633, 21634, 21635, 21636, 21637, 21638, 21639, 21640, 21641, 21642, ...
Number of generated alerts: 
      1 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 12240 packets: 11833, 11834, 11835, 11836, 11837, 11838, 11839, 11840, 11841, 11842, 11843, 11844, 11845, 11846, 11847, 11848, 11849, 11850, ...
      2 in 6528 packets: 21625, 21626, 21627, 21628, 21629, 21630, 21631, 21632, 21633, 21634, 21635, 21636, 21637, 21638, 21639, 21640, 21641, 21642, ...
Survey memory error: 
      0 in 31008 packets (all packets)
Event memory error: 
      0 in 31008 packets (all packets)
Configuration error: 
      0 in 31008 packets (all packets)
SDRAM CRC error: 
      0 in 31008 packets (all packets)
Radio gain: 
      0 in 31008 packets (all packets)
Radio bandwidth: 
      0 in 31008 packets (all packets)
Digital gain: 
      0 in 31008 packets (all packets)
Analog gain: 
      0 in 31008 packets (all packets)

EVENT DATA READER
-- File 20220703_081058_1229.dat, 152 EW pages found. Extra packets: 0.
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      1 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 12240 packets: 11833, 11834, 11835, 11836, 11837, 11838, 11839, 11840, 11841, 11842, 11843, 11844, 11845, 11846, 11847, 11848, 11849, 11850, ...
      2 in 6528 packets: 21625, 21626, 21627, 21628, 21629, 21630, 21631, 21632, 21633, 21634, 21635, 21636, 21637, 21638, 21639, 21640, 21641, 21642, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 11.0056241 to 33.6030951
      Difference between the neighboring packets: 
      0.0000170 in 31004 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      22.2229610 in 1 packets: 11832
      -0.3190390 in 1 packets: 21624
      -22.4310070 in 1 packets: 30600
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 30856 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 150 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      87 in 1 packets: 21624
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 30856 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 147 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -47 in 1 packets: 11832
      -59 in 1 packets: 21624
      -31 in 1 packets: 28152
      -11 in 1 packets: 30600
   MIU event flag 
      1 in 31008 packets (all packets)
   Time triggered event flag 
      0 in 31008 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-07-03T08:11:10.241.434.620
   Min EW header abs. time found among all events            : 2022-07-03T08:11:10.296.923.000
   Max EW event (1st sample) abs. time found among all events: 2022-07-03T08:11:32.630.922.620
   Max EW header abs. time found among all events            : 2022-07-03T08:11:32.686.411.000
   TM values from -296 to 248 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -296 to 241 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=9 C1MHz=0.296923
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 11.0056241
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-07-03T08:11:10.296.923.000
   Event 1st sample abs. time : 2022-07-03T08:11:10.241.434.620)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.005624100 corrected by 40bit 10MHz counter of 10.764188600 obtained at 1Hz counter of 9. Result: 9.241436393
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):               9.241436218
   Event data descriptor XO time (reconstructed seconds from XP start)    :               9.241436393
   Sampling frequency correction factor for this event:     0.999996300

 Event no. 2 total number of packets: 6528
    Page no. 1:  204 packets  -- OK;  memory page(s):
      192 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      193 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      194 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      195 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      196 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      197 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      198 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      199 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      200 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      216 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      217 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      218 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      219 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      220 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 32  ordered packets: 6528
   TM values from -282 to 244 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=31 C1MHz=0.575435
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 6527 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 33.2840561
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-07-03T08:11:32.575.435.000
   Event 1st sample abs. time : 2022-07-03T08:11:32.519.946.620)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 31, 32bit 10MHz counter = 33.284056100 corrected by 40bit 10MHz counter of 29.764119500 obtained at 1Hz counter of 28. Result: 31.519949624
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              31.519949449
   Event data descriptor XO time (reconstructed seconds from XP start)    :              31.519949624
   Sampling frequency correction factor for this event:     0.999996300

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -277 to 248 median: -5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=31 C1MHz=0.686411
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 33.3950321
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-07-03T08:11:32.686.411.000
   Event 1st sample abs. time : 2022-07-03T08:11:32.630.922.620)
   --- Rel. time correction for 1 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 31, 32bit 10MHz counter = 33.395032100 corrected by 40bit 10MHz counter of 29.764119500 obtained at 1Hz counter of 28. Result: 31.630926034
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              31.630925859
   Event data descriptor XO time (reconstructed seconds from XP start)    :              31.630926034
   Sampling frequency correction factor for this event:     0.999996300
--
Extremes in separate events:

 - Event number: 1 from 2022-07-03T08:11:10.241.434.620 to 2022-07-03T08:11:10.449.515.389 (XO: 9.241436218 - 9.449516988)
   Min TM value:-296 at an offset of 53160.659 usec -> 2022-07-03T08:11:10.294.595.279 (XO: 9.294596878)
   Max TM value:241 at an offset of 133061.192 usec -> 2022-07-03T08:11:10.374.495.812 (XO: 9.374497411)
   Max dTM/dt [1/us]:2.52e+03 at an offset of 53160.772 usec -> 2022-07-03T08:11:10.294.595.391 (XO: 9.294596990)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 53158.309 usec -> 2022-07-03T08:11:10.294.592.929 (XO: 9.294594527)
   Med: -5.00000   Ave: -4.5189388   Std: 18.335816

 - Event number: 2 from 2022-07-03T08:11:32.519.946.620 to 2022-07-03T08:11:32.630.923.030 (XO: 31.519949449 - 31.630925859)
   Min TM value:-282 at an offset of 53629.298 usec -> 2022-07-03T08:11:32.573.575.918 (XO: 31.573578747)
   Max TM value:244 at an offset of 53626.898 usec -> 2022-07-03T08:11:32.573.573.518 (XO: 31.573576347)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 53629.411 usec -> 2022-07-03T08:11:32.573.576.030 (XO: 31.573578860)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 53626.998 usec -> 2022-07-03T08:11:32.573.573.618 (XO: 31.573576447)
   Med: -5.00000   Ave: -4.5221480   Std: 10.939742

 - Event number: 3 from 2022-07-03T08:11:32.630.922.620 to 2022-07-03T08:11:32.839.003.389 (XO: 31.630925859 - 31.839006629)
   Min TM value:-277 at an offset of 102702.692 usec -> 2022-07-03T08:11:32.733.625.312 (XO: 31.733628552)
   Max TM value:248 at an offset of 52655.407 usec -> 2022-07-03T08:11:32.683.578.027 (XO: 31.683581267)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 142690.053 usec -> 2022-07-03T08:11:32.773.612.672 (XO: 31.773615912)
   Min dTM/dt [1/us]:-2.71e+03 at an offset of 102702.567 usec -> 2022-07-03T08:11:32.733.625.187 (XO: 31.733628427)
   Med: -5.00000   Ave: -4.5388131   Std: 10.881935
