-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 17:10:00 2023 UTC
-- File 20220630_164823_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.366113 to 87.376537
-each separately from 2s to 87s and from 130126us to 774411us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.763724 in 1 packets: 18
      19.443380 in 1 packets: 36
      6.200038 in 1 packets: 54
      25.761021 in 1 packets: 72
      12.840527 in 1 packets: 90
MIU packet time from 2022-06-30T16:48:28.366.113.000 to 2022-06-30T16:49:53.376.537.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.763724 in 1 packets: 18
      19.443380 in 1 packets: 36
      6.200038 in 1 packets: 54
      25.761021 in 1 packets: 72
      12.840527 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.1145405 to 89.1142293
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999230 in 1 packets: 18
      18.9999304 in 1 packets: 36
      5.9999781 in 1 packets: 54
      25.9999050 in 1 packets: 72
      12.9999523 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220630_164823_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.4806541 to 4.4809431
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7636470  -- OK for SL --
   MIU time from 2022-06-30T16:48:28.366.113.000 to 2022-06-30T16:48:28.366.402.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.763724

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2445901 to 25.2448791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4433110  -- OK for SL --
   MIU time from 2022-06-30T16:48:49.130.126.000 to 2022-06-30T16:48:49.130.415.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.443380

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.6881901 to 44.6884791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.2000150  -- OK for SL --
   MIU time from 2022-06-30T16:49:08.573.795.000 to 2022-06-30T16:49:08.574.084.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.200038

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.8884941 to 50.8887831
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.7609270  -- OK for SL --
   MIU time from 2022-06-30T16:49:14.774.122.000 to 2022-06-30T16:49:14.774.411.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.761021

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.6497101 to 76.6499991
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8404790  -- OK for SL --
   MIU time from 2022-06-30T16:49:40.535.432.000 to 2022-06-30T16:49:40.535.721.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.840527

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 89.4904781 to 89.4907671
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-30T16:49:53.376.248.000 to 2022-06-30T16:49:53.376.537.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.480654100 corrected by 40bit 10MHz counter of 4.114540500 obtained at 1Hz counter of 2. Result: 2.366114955
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-06-30T16:48:28.365.972.725
   Max 1st sample MIU time among all SL packets: 2022-06-30T16:49:53.376.396.725
   Min 1st sample XO time among all SL packets: 2.365974679
   Max 1st sample XO time among all SL packets: 87.376398918
   TM values from -284 to 250 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-30T16:48:28.365.972.725 (XO: 2.365974679 - 2.366280680, t=37.2 degC)  Rank: 00079
   TM values: Max =   226  Min =  -282  Ave =    -4.39  Std =    30.42
   Min TM value:-282 at an offset of 98.988 usec -> 2022-06-30T16:48:28.366.071.712 (XO: 2.366073667)
   Max TM value:226 at an offset of 99.275 usec -> 2022-06-30T16:48:28.366.072.000 (XO: 2.366073954)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 156.838 usec -> 2022-06-30T16:48:28.366.129.563 (XO: 2.366131517)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 96.688 usec -> 2022-06-30T16:48:28.366.069.412 (XO: 2.366071367)

 - Block number: 2 starting at 2022-06-30T16:48:49.129.985.725 (XO: 23.129986806 - 23.130292807, t=37.2 degC)  Rank: 00076
   TM values: Max =   232  Min =  -279  Ave =    -4.39  Std =    28.99
   Min TM value:-279 at an offset of 110.938 usec -> 2022-06-30T16:48:49.130.096.662 (XO: 23.130097744)
   Max TM value:232 at an offset of 108.475 usec -> 2022-06-30T16:48:49.130.094.200 (XO: 23.130095281)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 111.050 usec -> 2022-06-30T16:48:49.130.096.775 (XO: 23.130097856)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 108.600 usec -> 2022-06-30T16:48:49.130.094.325 (XO: 23.130095406)

 - Block number: 3 starting at 2022-06-30T16:49:08.573.654.725 (XO: 42.573658048 - 42.573964049, t=37.2 degC)  Rank: 00078
   TM values: Max =   180  Min =  -238  Ave =    -4.46  Std =    22.26
   Min TM value:-238 at an offset of 126.163 usec -> 2022-06-30T16:49:08.573.780.887 (XO: 42.573784211)
   Max TM value:180 at an offset of 123.700 usec -> 2022-06-30T16:49:08.573.778.425 (XO: 42.573781748)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 126.300 usec -> 2022-06-30T16:49:08.573.781.025 (XO: 42.573784348)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 123.813 usec -> 2022-06-30T16:49:08.573.778.537 (XO: 42.573781860)

 - Block number: 4 starting at 2022-06-30T16:49:14.773.981.725 (XO: 48.773984689 - 48.774290690, t=37.2 degC)  Rank: 00089
   TM values: Max =   195  Min =  -284  Ave =    -4.49  Std =    25.90
   Min TM value:-284 at an offset of 123.363 usec -> 2022-06-30T16:49:14.774.105.087 (XO: 48.774108052)
   Max TM value:195 at an offset of 120.888 usec -> 2022-06-30T16:49:14.774.102.612 (XO: 48.774105577)
   Max dTM/dt [1/us]:2.40e+03 at an offset of 123.475 usec -> 2022-06-30T16:49:14.774.105.200 (XO: 48.774108164)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 121.013 usec -> 2022-06-30T16:49:14.774.102.737 (XO: 48.774105702)

 - Block number: 5 starting at 2022-06-30T16:49:40.535.291.725 (XO: 74.535294806 - 74.535600807, t=37.2 degC)  Rank: 00079
   TM values: Max =   250  Min =  -252  Ave =    -4.52  Std =    23.58
   Min TM value:-252 at an offset of 127.088 usec -> 2022-06-30T16:49:40.535.418.812 (XO: 74.535421894)
   Max TM value:250 at an offset of 124.638 usec -> 2022-06-30T16:49:40.535.416.362 (XO: 74.535419444)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 124.575 usec -> 2022-06-30T16:49:40.535.416.300 (XO: 74.535419381)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 124.750 usec -> 2022-06-30T16:49:40.535.416.475 (XO: 74.535419556)

 - Block number: 6 starting at 2022-06-30T16:49:53.376.107.725 (XO: 87.376109917 - 87.376415918, t=37.2 degC)  Rank: 00081
   TM values: Max =   230  Min =  -240  Ave =    -4.48  Std =    27.48
   Min TM value:-240 at an offset of 112.025 usec -> 2022-06-30T16:49:53.376.219.750 (XO: 87.376221942)
   Max TM value:230 at an offset of 111.813 usec -> 2022-06-30T16:49:53.376.219.537 (XO: 87.376221730)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 155.338 usec -> 2022-06-30T16:49:53.376.263.063 (XO: 87.376265255)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 111.913 usec -> 2022-06-30T16:49:53.376.219.637 (XO: 87.376221830)
