-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar 22 12:18:22 2023 UTC
-- File 20220630_125906_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 15.006238 to 15.406559
-each separately from 15s to 15s and from 6238us to 406559us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.399743 in 1 packets: 18
MIU packet time from 2022-06-30T12:59:24.006.238.000 to 2022-06-30T12:59:24.406.559.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      0.399743 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.7648708 to 16.7648708
   Differences in neigboring packets: 
      0.0000000 in 35 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220630_125906_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7711100 to 16.7713990
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.3997430  -- OK for SL --
   MIU time from 2022-06-30T12:59:24.006.238.000 to 2022-06-30T12:59:24.006.527.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.399743

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1711420 to 17.1714310
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-30T12:59:24.406.270.000 to 2022-06-30T12:59:24.406.559.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 15, 32bit 10MHz counter = 16.771110000 corrected by 40bit 10MHz counter of 16.764870800 obtained at 1Hz counter of 15. Result: 15.006239222
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-06-30T12:59:24.006.097.725
   Max 1st sample MIU time among all SL packets: 2022-06-30T12:59:24.406.418.725
   Min 1st sample XO time among all SL packets: 15.006098947
   Max 1st sample XO time among all SL packets: 15.406421388
   TM values from -248 to 235 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-30T12:59:24.006.097.725 (XO: 15.006098947 - 15.006404948, t=37.2 degC)  Rank: 00092
   TM values: Max =   235  Min =  -248  Ave =    -4.54  Std =    21.29
   Min TM value:-248 at an offset of 127.088 usec -> 2022-06-30T12:59:24.006.224.812 (XO: 15.006226035)
   Max TM value:235 at an offset of 124.650 usec -> 2022-06-30T12:59:24.006.222.375 (XO: 15.006223597)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 127.200 usec -> 2022-06-30T12:59:24.006.224.925 (XO: 15.006226147)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 124.750 usec -> 2022-06-30T12:59:24.006.222.475 (XO: 15.006223697)

 - Block number: 2 starting at 2022-06-30T12:59:24.406.129.725 (XO: 15.406132387 - 15.406438388, t=37.2 degC)  Rank: 00077
   TM values: Max =   184  Min =  -232  Ave =    -4.52  Std =    30.95
   Min TM value:-232 at an offset of 102.675 usec -> 2022-06-30T12:59:24.406.232.400 (XO: 15.406235062)
   Max TM value:184 at an offset of 100.175 usec -> 2022-06-30T12:59:24.406.229.900 (XO: 15.406232562)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 102.788 usec -> 2022-06-30T12:59:24.406.232.512 (XO: 15.406235175)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 100.288 usec -> 2022-06-30T12:59:24.406.230.012 (XO: 15.406232675)
