-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 16 20:29:42 2023 UTC
-- File 20220611_134603_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.138913 to 70.421565
-each separately from 11s to 70s and from 138913us to 598457us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.458966 in 1 packets: 18
      8.803998 in 1 packets: 36
      21.106487 in 1 packets: 54
      12.911756 in 1 packets: 72
MIU packet time from 2022-06-11T13:46:17.138.913.000 to 2022-06-11T13:47:16.421.565.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.458966 in 1 packets: 18
      8.803998 in 1 packets: 36
      21.106487 in 1 packets: 54
      12.911756 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.7644866 to 71.7642834
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999448 in 1 packets: 18
      8.9999690 in 1 packets: 36
      20.9999278 in 1 packets: 54
      12.9999552 in 1 packets: 72
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FF -> 56.3 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C5 -> 43.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 89 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 89 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220611_134603_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.9034010 to 12.9036900
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4589110  -- OK for SL --
   MIU time from 2022-06-11T13:46:17.138.913.000 to 2022-06-11T13:46:17.139.202.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.458966

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.3626010 to 29.3628900
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.8039670  -- OK for SL --
   MIU time from 2022-06-11T13:46:33.598.168.000 to 2022-06-11T13:46:33.598.457.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.803998

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.1668570 to 38.1671460
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.1064150  -- OK for SL --
   MIU time from 2022-06-11T13:46:42.402.455.000 to 2022-06-11T13:46:42.402.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.106487

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.2735610 to 59.2738500
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9117110  -- OK for SL --
   MIU time from 2022-06-11T13:47:03.509.231.000 to 2022-06-11T13:47:03.509.520.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.911756

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 72.1855610 to 72.1858500
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-11T13:47:16.421.276.000 to 2022-06-11T13:47:16.421.565.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.903401000 corrected by 40bit 10MHz counter of 12.764486600 obtained at 1Hz counter of 11. Result: 11.138914886
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-06-11T13:46:17.138.772.725
   Max 1st sample MIU time among all SL packets: 2022-06-11T13:47:16.421.424.725
   Min 1st sample XO time among all SL packets: 11.138774611
   Max 1st sample XO time among all SL packets: 70.421427758
   TM values from -300 to 230 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-11T13:46:17.138.772.725 (XO: 11.138774611 - 11.139080612, t=43.6 degC)  Rank: 00088
   TM values: Max =   227  Min =  -272  Ave =    -4.62  Std =    32.66
   Min TM value:-272 at an offset of 200.813 usec -> 2022-06-11T13:46:17.138.973.538 (XO: 11.138975424)
   Max TM value:227 at an offset of 203.138 usec -> 2022-06-11T13:46:17.138.975.863 (XO: 11.138977749)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 174.613 usec -> 2022-06-11T13:46:17.138.947.338 (XO: 11.138949224)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 200.713 usec -> 2022-06-11T13:46:17.138.973.438 (XO: 11.138975324)

 - Block number: 2 starting at 2022-06-11T13:46:33.598.027.725 (XO: 27.598031358 - 27.598337359, t=43.6 degC)  Rank: 00078
   TM values: Max =   204  Min =  -216  Ave =    -4.47  Std =    21.67
   Min TM value:-216 at an offset of 118.313 usec -> 2022-06-11T13:46:33.598.146.037 (XO: 27.598149671)
   Max TM value:204 at an offset of 118.050 usec -> 2022-06-11T13:46:33.598.145.775 (XO: 27.598149409)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 117.975 usec -> 2022-06-11T13:46:33.598.145.700 (XO: 27.598149334)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 118.150 usec -> 2022-06-11T13:46:33.598.145.875 (XO: 27.598149509)

 - Block number: 3 starting at 2022-06-11T13:46:42.402.314.725 (XO: 36.402317693 - 36.402623694, t=43.6 degC)  Rank: 00078
   TM values: Max =   230  Min =  -300  Ave =    -4.56  Std =    24.66
   Min TM value:-300 at an offset of 154.438 usec -> 2022-06-11T13:46:42.402.469.163 (XO: 36.402472131)
   Max TM value:230 at an offset of 112.488 usec -> 2022-06-11T13:46:42.402.427.212 (XO: 36.402430181)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 154.513 usec -> 2022-06-11T13:46:42.402.469.238 (XO: 36.402472206)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 112.588 usec -> 2022-06-11T13:46:42.402.427.312 (XO: 36.402430281)

 - Block number: 4 starting at 2022-06-11T13:47:03.509.090.725 (XO: 57.509094256 - 57.509400257, t=43.6 degC)  Rank: 00080
   TM values: Max =   228  Min =  -271  Ave =    -4.46  Std =    31.52
   Min TM value:-271 at an offset of 85.563 usec -> 2022-06-11T13:47:03.509.176.287 (XO: 57.509179819)
   Max TM value:228 at an offset of 156.376 usec -> 2022-06-11T13:47:03.509.247.100 (XO: 57.509250631)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 156.251 usec -> 2022-06-11T13:47:03.509.246.975 (XO: 57.509250506)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 85.388 usec -> 2022-06-11T13:47:03.509.176.112 (XO: 57.509179644)

 - Block number: 5 starting at 2022-06-11T13:47:16.421.135.725 (XO: 70.421138757 - 70.421444758, t=43.6 degC)  Rank: 00080
   TM values: Max =   206  Min =  -250  Ave =    -4.47  Std =    30.32
   Min TM value:-250 at an offset of 215.226 usec -> 2022-06-11T13:47:16.421.350.950 (XO: 70.421353983)
   Max TM value:206 at an offset of 217.538 usec -> 2022-06-11T13:47:16.421.353.263 (XO: 70.421356295)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 217.376 usec -> 2022-06-11T13:47:16.421.353.100 (XO: 70.421356133)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 133.125 usec -> 2022-06-11T13:47:16.421.268.850 (XO: 70.421271882)
