-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar 13 12:32:16 2023 UTC
-- File 20220602_132127_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.520067 to 36.200702
-each separately from 4s to 36s and from 200413us to 520356us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.689873 in 1 packets: 18
      16.989895 in 1 packets: 36
MIU packet time from 2022-06-02T13:21:34.520.067.000 to 2022-06-02T13:22:06.200.702.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.689873 in 1 packets: 18
      16.989895 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.7595905 to 37.7594845
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999503 in 1 packets: 18
      16.9999437 in 1 packets: 36
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C5 -> 43.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220602_132127_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.2796593 to 6.2799483
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6898230  -- OK for SL --
   MIU time from 2022-06-02T13:21:34.520.067.000 to 2022-06-02T13:21:34.520.356.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.689873

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.9697713 to 20.9700603
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9898390  -- OK for SL --
   MIU time from 2022-06-02T13:21:49.210.229.000 to 2022-06-02T13:21:49.210.518.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.989895

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9598993 to 37.9601883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-06-02T13:22:06.200.413.000 to 2022-06-02T13:22:06.200.702.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.279659300 corrected by 40bit 10MHz counter of 5.759590500 obtained at 1Hz counter of 4. Result: 4.520070568
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-06-02T13:21:34.519.926.725
   Max 1st sample MIU time among all SL packets: 2022-06-02T13:22:06.200.561.725
   Min 1st sample XO time among all SL packets: 4.519930293
   Max 1st sample XO time among all SL packets: 36.200564187
   TM values from -275 to 242 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-06-02T13:21:34.519.926.725 (XO: 4.519930293 - 4.520236294, t=43.6 degC)  Rank: 00083
   TM values: Max =   232  Min =  -245  Ave =    -4.43  Std =    29.68
   Min TM value:-245 at an offset of 149.938 usec -> 2022-06-02T13:21:34.520.076.663 (XO: 4.520080231)
   Max TM value:232 at an offset of 152.313 usec -> 2022-06-02T13:21:34.520.079.038 (XO: 4.520082606)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 123.613 usec -> 2022-06-02T13:21:34.520.050.337 (XO: 4.520053906)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 126.050 usec -> 2022-06-02T13:21:34.520.052.775 (XO: 4.520056343)

 - Block number: 2 starting at 2022-06-02T13:21:49.210.088.725 (XO: 19.210090918 - 19.210396919, t=43.6 degC)  Rank: 00078
   TM values: Max =   233  Min =  -225  Ave =    -4.35  Std =    36.41
   Min TM value:-225 at an offset of 123.050 usec -> 2022-06-02T13:21:49.210.211.775 (XO: 19.210213969)
   Max TM value:233 at an offset of 123.338 usec -> 2022-06-02T13:21:49.210.212.062 (XO: 19.210214256)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 123.200 usec -> 2022-06-02T13:21:49.210.211.925 (XO: 19.210214119)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 227.426 usec -> 2022-06-02T13:21:49.210.316.150 (XO: 19.210318344)

 - Block number: 3 starting at 2022-06-02T13:22:06.200.272.725 (XO: 36.200275186 - 36.200581187, t=43.6 degC)  Rank: 00083
   TM values: Max =   242  Min =  -275  Ave =    -4.40  Std =    32.90
   Min TM value:-275 at an offset of 204.813 usec -> 2022-06-02T13:22:06.200.477.538 (XO: 36.200479999)
   Max TM value:242 at an offset of 207.176 usec -> 2022-06-02T13:22:06.200.479.900 (XO: 36.200482362)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 207.076 usec -> 2022-06-02T13:22:06.200.479.800 (XO: 36.200482262)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 181.588 usec -> 2022-06-02T13:22:06.200.454.313 (XO: 36.200456774)
