-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 21:40:48 2023 UTC
-- File 20220118_053110_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.818240 to 23.118846
-each separately from 1s to 23s and from 118557us to 818529us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.300028 in 1 packets: 18
MIU packet time from 2022-01-18T05:31:14.818.240.000 to 2022-01-18T05:31:36.118.846.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.300028 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7781301 to 24.7780522
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999221 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220118_053110_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      52 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.5963701 to 3.5966591
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2999510  -- OK for SL --
   MIU time from 2022-01-18T05:31:14.818.240.000 to 2022-01-18T05:31:14.818.529.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.300028

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      53 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.8966101 to 24.8968991
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-18T05:31:36.118.557.000 to 2022-01-18T05:31:36.118.846.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.596370100 corrected by 40bit 10MHz counter of 2.778130100 obtained at 1Hz counter of 1. Result: 1.818242946
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-01-18T05:31:14.818.099.725
   Max 1st sample MIU time among all SL packets: 2022-01-18T05:31:36.118.705.725
   Min 1st sample XO time among all SL packets: 1.818102670
   Max 1st sample XO time among all SL packets: 23.118707040
   TM values from -177 to 144 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-18T05:31:14.818.099.725 (XO: 1.818102670 - 1.818408671, t=37.2 degC)  Rank: 00052
   TM values: Max =   144  Min =  -170  Ave =    -4.51  Std =    19.53
   Min TM value:-170 at an offset of 185.063 usec -> 2022-01-18T05:31:14.818.284.788 (XO: 1.818287733)
   Max TM value:144 at an offset of 124.988 usec -> 2022-01-18T05:31:14.818.224.712 (XO: 1.818227658)
   Max dTM/dt [1/us]:1.17e+03 at an offset of 127.538 usec -> 2022-01-18T05:31:14.818.227.262 (XO: 1.818230208)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 125.100 usec -> 2022-01-18T05:31:14.818.224.825 (XO: 1.818227771)

 - Block number: 2 starting at 2022-01-18T05:31:36.118.416.725 (XO: 23.118418039 - 23.118724041, t=37.2 degC)  Rank: 00053
   TM values: Max =   143  Min =  -177  Ave =    -4.67  Std =    18.95
   Min TM value:-177 at an offset of 156.013 usec -> 2022-01-18T05:31:36.118.572.738 (XO: 23.118574053)
   Max TM value:143 at an offset of 115.700 usec -> 2022-01-18T05:31:36.118.532.425 (XO: 23.118533740)
   Max dTM/dt [1/us]:1.33e+03 at an offset of 158.301 usec -> 2022-01-18T05:31:36.118.575.025 (XO: 23.118576340)
   Min dTM/dt [1/us]:-1.40e+03 at an offset of 113.138 usec -> 2022-01-18T05:31:36.118.529.862 (XO: 23.118531177)
