-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 31 09:18:22 2022 UTC
-- File 20211006_222824_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.573062 to 20.704332
-each separately from 4s to 20s and from 573062us to 704332us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.130692 in 1 packets: 18
MIU packet time from 2021-10-06T22:28:31.573.062.000 to 2021-10-06T22:28:47.704.332.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.130692 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8129981 to 21.8129446
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999465 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211006_222824_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.3860607 to 6.3863497
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1306390  -- OK for SL --
   MIU time from 2021-10-06T22:28:31.573.062.000 to 2021-10-06T22:28:31.573.351.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.130692

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.5169887 to 22.5172777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-06T22:28:47.704.043.000 to 2021-10-06T22:28:47.704.332.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.386060700 corrected by 40bit 10MHz counter of 5.812998100 obtained at 1Hz counter of 4. Result: 4.573064491
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-06T22:28:31.572.921.725
   Max 1st sample MIU time among all SL packets: 2021-10-06T22:28:47.704.191.725
   Min 1st sample XO time among all SL packets: 4.572924216
   Max 1st sample XO time among all SL packets: 20.704195149
   TM values from -262 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-06T22:28:31.572.921.725 (XO: 4.572924216 - 4.573230217, t=37.2 degC)  Rank: 00081
   TM values: Max =   230  Min =  -250  Ave =    -4.53  Std =    20.43
   Min TM value:-250 at an offset of 113.013 usec -> 2021-10-06T22:28:31.573.034.737 (XO: 4.573037229)
   Max TM value:230 at an offset of 112.738 usec -> 2021-10-06T22:28:31.573.034.462 (XO: 4.573036954)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 115.425 usec -> 2021-10-06T22:28:31.573.037.150 (XO: 4.573039641)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 112.875 usec -> 2021-10-06T22:28:31.573.034.600 (XO: 4.573037091)

 - Block number: 2 starting at 2021-10-06T22:28:47.703.902.725 (XO: 20.703906148 - 20.704212149, t=37.2 degC)  Rank: 00078
   TM values: Max =   240  Min =  -262  Ave =    -4.38  Std =    28.84
   Min TM value:-262 at an offset of 130.925 usec -> 2021-10-06T22:28:47.704.033.650 (XO: 20.704037073)
   Max TM value:240 at an offset of 89.900 usec -> 2021-10-06T22:28:47.703.992.625 (XO: 20.703996048)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 92.575 usec -> 2021-10-06T22:28:47.703.995.300 (XO: 20.703998723)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 90.038 usec -> 2021-10-06T22:28:47.703.992.762 (XO: 20.703996186)
