-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 31 07:43:22 2022 UTC
-- File 20211006_203229_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.013523 to 87.313295
-each separately from 12s to 87s and from 13523us to 924415us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.509779 in 1 packets: 18
      21.400246 in 1 packets: 36
      5.699875 in 1 packets: 54
      16.699448 in 1 packets: 72
      24.988690 in 1 packets: 90
MIU packet time from 2021-10-06T20:32:44.013.523.000 to 2021-10-06T20:33:59.313.295.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.509779 in 1 packets: 18
      21.400246 in 1 packets: 36
      5.699875 in 1 packets: 54
      16.699448 in 1 packets: 72
      24.988690 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7134234 to 88.7131725
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999800 in 1 packets: 18
      20.9999297 in 1 packets: 36
      5.9999799 in 1 packets: 54
      16.9999432 in 1 packets: 72
      24.9999163 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 71 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 71 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20211006_203229_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.7269477 to 13.7272367
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.5097590  -- OK for SL --
   MIU time from 2021-10-06T20:32:44.013.523.000 to 2021-10-06T20:32:44.013.812.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.509779

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2369957 to 20.2372847
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.4001750  -- OK for SL --
   MIU time from 2021-10-06T20:32:50.523.591.000 to 2021-10-06T20:32:50.523.880.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.400246

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.6374597 to 41.6377487
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.6998550  -- OK for SL --
   MIU time from 2021-10-06T20:33:11.924.126.000 to 2021-10-06T20:33:11.924.415.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.699875

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.3376037 to 47.3378927
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6993910  -- OK for SL --
   MIU time from 2021-10-06T20:33:17.624.290.000 to 2021-10-06T20:33:17.624.579.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.699448

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.0372837 to 64.0375727
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9886070  -- OK for SL --
   MIU time from 2021-10-06T20:33:34.324.027.000 to 2021-10-06T20:33:34.324.316.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.988690

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 89.0261797 to 89.0264687
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-06T20:33:59.313.006.000 to 2021-10-06T20:33:59.313.295.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.726947700 corrected by 40bit 10MHz counter of 13.713423400 obtained at 1Hz counter of 12. Result: 12.013524346
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-06T20:32:44.013.382.725
   Max 1st sample MIU time among all SL packets: 2021-10-06T20:33:59.313.154.725
   Min 1st sample XO time among all SL packets: 12.013384071
   Max 1st sample XO time among all SL packets: 87.313156958
   TM values from -282 to 248 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-06T20:32:44.013.382.725 (XO: 12.013384071 - 12.013690072, t=37.2 degC)  Rank: 00079
   TM values: Max =   203  Min =  -254  Ave =    -4.43  Std =    19.14
   Min TM value:-254 at an offset of 115.250 usec -> 2021-10-06T20:32:44.013.497.975 (XO: 12.013499321)
   Max TM value:203 at an offset of 117.825 usec -> 2021-10-06T20:32:44.013.500.550 (XO: 12.013501896)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 117.638 usec -> 2021-10-06T20:32:44.013.500.362 (XO: 12.013501708)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 115.163 usec -> 2021-10-06T20:32:44.013.497.887 (XO: 12.013499233)

 - Block number: 2 starting at 2021-10-06T20:32:50.523.450.725 (XO: 18.523453805 - 18.523759806, t=37.2 degC)  Rank: 00085
   TM values: Max =   211  Min =  -279  Ave =    -4.46  Std =    29.70
   Min TM value:-279 at an offset of 159.301 usec -> 2021-10-06T20:32:50.523.610.025 (XO: 18.523613105)
   Max TM value:211 at an offset of 131.100 usec -> 2021-10-06T20:32:50.523.581.825 (XO: 18.523584905)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 176.113 usec -> 2021-10-06T20:32:50.523.626.838 (XO: 18.523629918)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 128.275 usec -> 2021-10-06T20:32:50.523.579.000 (XO: 18.523582080)

 - Block number: 3 starting at 2021-10-06T20:33:11.923.985.725 (XO: 39.923989467 - 39.924295468, t=37.2 degC)  Rank: 00082
   TM values: Max =   233  Min =  -269  Ave =    -4.34  Std =    27.92
   Min TM value:-269 at an offset of 146.325 usec -> 2021-10-06T20:33:11.924.132.050 (XO: 39.924135792)
   Max TM value:233 at an offset of 117.988 usec -> 2021-10-06T20:33:11.924.103.712 (XO: 39.924107454)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 148.676 usec -> 2021-10-06T20:33:11.924.134.400 (XO: 39.924138142)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 146.125 usec -> 2021-10-06T20:33:11.924.131.850 (XO: 39.924135592)

 - Block number: 4 starting at 2021-10-06T20:33:17.624.149.725 (XO: 45.624152485 - 45.624458486, t=37.2 degC)  Rank: 00079
   TM values: Max =   248  Min =  -271  Ave =    -4.38  Std =    26.00
   Min TM value:-271 at an offset of 103.000 usec -> 2021-10-06T20:33:17.624.252.725 (XO: 45.624255485)
   Max TM value:248 at an offset of 102.738 usec -> 2021-10-06T20:33:17.624.252.462 (XO: 45.624255223)
   Max dTM/dt [1/us]:2.42e+03 at an offset of 105.375 usec -> 2021-10-06T20:33:17.624.255.100 (XO: 45.624257860)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 102.863 usec -> 2021-10-06T20:33:17.624.252.587 (XO: 45.624255348)

 - Block number: 5 starting at 2021-10-06T20:33:34.323.886.725 (XO: 62.323888326 - 62.324194327, t=37.2 degC)  Rank: 00079
   TM values: Max =   219  Min =  -282  Ave =    -4.42  Std =    24.18
   Min TM value:-282 at an offset of 129.788 usec -> 2021-10-06T20:33:34.324.016.512 (XO: 62.324018114)
   Max TM value:219 at an offset of 132.313 usec -> 2021-10-06T20:33:34.324.019.037 (XO: 62.324020639)
   Max dTM/dt [1/us]:2.63e+03 at an offset of 132.138 usec -> 2021-10-06T20:33:34.324.018.862 (XO: 62.324020464)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 129.575 usec -> 2021-10-06T20:33:34.324.016.300 (XO: 62.324017902)

 - Block number: 6 starting at 2021-10-06T20:33:59.312.865.725 (XO: 87.312867957 - 87.313173958, t=37.2 degC)  Rank: 00075
   TM values: Max =   224  Min =  -250  Ave =    -4.47  Std =    22.10
   Min TM value:-250 at an offset of 128.375 usec -> 2021-10-06T20:33:59.312.994.100 (XO: 87.312996333)
   Max TM value:224 at an offset of 128.100 usec -> 2021-10-06T20:33:59.312.993.825 (XO: 87.312996058)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 130.725 usec -> 2021-10-06T20:33:59.312.996.450 (XO: 87.312998683)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 128.213 usec -> 2021-10-06T20:33:59.312.993.937 (XO: 87.312996170)
