-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 31 04:09:03 2022 UTC
-- File 20211006_151133_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.261721 to 20.052398
-each separately from 9s to 20s and from 52109us to 262010us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.790099 in 1 packets: 18
MIU packet time from 2021-10-06T15:11:45.261.721.000 to 2021-10-06T15:11:56.052.398.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.790099 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8621572 to 21.8621203
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999631 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211006_151133_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1238788 to 11.1241678
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7900630  -- OK for SL --
   MIU time from 2021-10-06T15:11:45.261.721.000 to 2021-10-06T15:11:45.262.010.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.790099

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9142308 to 21.9145198
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-06T15:11:56.052.109.000 to 2021-10-06T15:11:56.052.398.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.123878800 corrected by 40bit 10MHz counter of 10.862157200 obtained at 1Hz counter of 9. Result: 9.261722490
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-06T15:11:45.261.580.725
   Max 1st sample MIU time among all SL packets: 2021-10-06T15:11:56.052.257.725
   Min 1st sample XO time among all SL packets: 9.261582214
   Max 1st sample XO time among all SL packets: 20.052259403
   TM values from -290 to 257 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-06T15:11:45.261.580.725 (XO: 9.261582214 - 9.261888215, t=37.2 degC)  Rank: 00079
   TM values: Max =   228  Min =  -243  Ave =    -4.54  Std =    20.47
   Min TM value:-243 at an offset of 109.788 usec -> 2021-10-06T15:11:45.261.690.512 (XO: 9.261692002)
   Max TM value:228 at an offset of 109.488 usec -> 2021-10-06T15:11:45.261.690.212 (XO: 9.261691702)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 112.200 usec -> 2021-10-06T15:11:45.261.692.925 (XO: 9.261694415)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 109.613 usec -> 2021-10-06T15:11:45.261.690.337 (XO: 9.261691827)

 - Block number: 2 starting at 2021-10-06T15:11:56.051.968.725 (XO: 20.051970402 - 20.052276403, t=37.2 degC)  Rank: 00091
   TM values: Max =   257  Min =  -290  Ave =    -4.51  Std =    33.09
   Min TM value:-290 at an offset of 192.213 usec -> 2021-10-06T15:11:56.052.160.938 (XO: 20.052162615)
   Max TM value:257 at an offset of 121.763 usec -> 2021-10-06T15:11:56.052.090.487 (XO: 20.052092165)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 124.438 usec -> 2021-10-06T15:11:56.052.093.162 (XO: 20.052094840)
   Min dTM/dt [1/us]:-2.70e+03 at an offset of 121.888 usec -> 2021-10-06T15:11:56.052.090.612 (XO: 20.052092290)
