-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 30 23:37:35 2022 UTC
-- File 20211006_101814_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.290870 to 29.149482
-each separately from 4s to 29s and from 149193us to 291159us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.858034 in 1 packets: 18
MIU packet time from 2021-10-06T10:18:21.290.870.000 to 2021-10-06T10:18:46.149.482.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.858034 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8603758 to 30.8602923
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999165 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211006_101814_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.1512468 to 6.1515358
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.8579510  -- OK for SL --
   MIU time from 2021-10-06T10:18:21.290.870.000 to 2021-10-06T10:18:21.291.159.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.858034

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.0094868 to 31.0097758
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-06T10:18:46.149.193.000 to 2021-10-06T10:18:46.149.482.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.151246800 corrected by 40bit 10MHz counter of 5.860375800 obtained at 1Hz counter of 4. Result: 4.290871960
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-06T10:18:21.290.729.725
   Max 1st sample MIU time among all SL packets: 2021-10-06T10:18:46.149.341.725
   Min 1st sample XO time among all SL packets: 4.290731684
   Max 1st sample XO time among all SL packets: 29.149343733
   TM values from -251 to 209 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-06T10:18:21.290.729.725 (XO: 4.290731684 - 4.291037685, t=37.2 degC)  Rank: 00079
   TM values: Max =   209  Min =  -251  Ave =    -4.32  Std =    19.30
   Min TM value:-251 at an offset of 122.475 usec -> 2021-10-06T10:18:21.290.852.200 (XO: 4.290854160)
   Max TM value:209 at an offset of 122.150 usec -> 2021-10-06T10:18:21.290.851.875 (XO: 4.290853835)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 124.800 usec -> 2021-10-06T10:18:21.290.854.525 (XO: 4.290856485)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 122.250 usec -> 2021-10-06T10:18:21.290.851.975 (XO: 4.290853935)

 - Block number: 2 starting at 2021-10-06T10:18:46.149.052.725 (XO: 29.149054732 - 29.149360733, t=37.2 degC)  Rank: 00077
   TM values: Max =   208  Min =  -234  Ave =    -4.30  Std =    23.25
   Min TM value:-234 at an offset of 120.413 usec -> 2021-10-06T10:18:46.149.173.137 (XO: 29.149175145)
   Max TM value:208 at an offset of 123.038 usec -> 2021-10-06T10:18:46.149.175.762 (XO: 29.149177770)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 122.888 usec -> 2021-10-06T10:18:46.149.175.612 (XO: 29.149177620)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 120.188 usec -> 2021-10-06T10:18:46.149.172.912 (XO: 29.149174920)
