-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 29 05:30:15 2022 UTC
-- File 20211004_071535_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.656201 to 87.703826
-each separately from 8s to 87s and from 309106us to 981416us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.088369 in 1 packets: 18
      7.634090 in 1 packets: 36
      22.929579 in 1 packets: 54
      6.671732 in 1 packets: 72
      26.722121 in 1 packets: 90
MIU packet time from 2021-10-04T07:15:46.656.201.000 to 2021-10-04T07:17:05.703.826.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.088369 in 1 packets: 18
      7.634090 in 1 packets: 36
      22.929579 in 1 packets: 54
      6.671732 in 1 packets: 72
      26.722121 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.7619232 to 88.7616587
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999497 in 1 packets: 18
      7.9999733 in 1 packets: 36
      22.9999231 in 1 packets: 54
      5.9999799 in 1 packets: 72
      26.9999095 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      3 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20211004_071535_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.4181247 to 10.4184137
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0883190  -- OK for SL --
   MIU time from 2021-10-04T07:15:46.656.201.000 to 2021-10-04T07:15:46.656.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.088369

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.5067327 to 25.5070217
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.6340630  -- OK for SL --
   MIU time from 2021-10-04T07:16:01.744.859.000 to 2021-10-04T07:16:01.745.148.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.634090

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.1410847 to 33.1413737
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.9295030  -- OK for SL --
   MIU time from 2021-10-04T07:16:09.379.238.000 to 2021-10-04T07:16:09.379.527.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.929579

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      99 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.0708767 to 56.0711657
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6717110  -- OK for SL --
   MIU time from 2021-10-04T07:16:32.309.106.000 to 2021-10-04T07:16:32.309.395.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.671732

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.7428767 to 62.7431657
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.7220310  -- OK for SL --
   MIU time from 2021-10-04T07:16:38.981.127.000 to 2021-10-04T07:16:38.981.416.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.722121

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 89.4651967 to 89.4654857
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-04T07:17:05.703.537.000 to 2021-10-04T07:17:05.703.826.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.418124700 corrected by 40bit 10MHz counter of 9.761923200 obtained at 1Hz counter of 8. Result: 8.656203665
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-04T07:15:46.656.060.725
   Max 1st sample MIU time among all SL packets: 2021-10-04T07:17:05.703.685.725
   Min 1st sample XO time among all SL packets: 8.656063390
   Max 1st sample XO time among all SL packets: 87.703689047
   TM values from -270 to 239 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-04T07:15:46.656.060.725 (XO: 8.656063390 - 8.656369391, t=37.2 degC)  Rank: 00074
   TM values: Max =   195  Min =  -233  Ave =    -4.50  Std =    20.56
   Min TM value:-233 at an offset of 119.588 usec -> 2021-10-04T07:15:46.656.180.312 (XO: 8.656182978)
   Max TM value:195 at an offset of 119.275 usec -> 2021-10-04T07:15:46.656.180.000 (XO: 8.656182665)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 121.925 usec -> 2021-10-04T07:15:46.656.182.650 (XO: 8.656185315)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 119.375 usec -> 2021-10-04T07:15:46.656.180.100 (XO: 8.656182765)

 - Block number: 2 starting at 2021-10-04T07:16:01.744.718.725 (XO: 23.744721983 - 23.745027984, t=37.2 degC)  Rank: 00084
   TM values: Max =   194  Min =  -255  Ave =    -4.48  Std =    21.54
   Min TM value:-255 at an offset of 118.888 usec -> 2021-10-04T07:16:01.744.837.612 (XO: 23.744840870)
   Max TM value:194 at an offset of 121.363 usec -> 2021-10-04T07:16:01.744.840.087 (XO: 23.744843345)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 121.213 usec -> 2021-10-04T07:16:01.744.839.937 (XO: 23.744843195)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 118.675 usec -> 2021-10-04T07:16:01.744.837.400 (XO: 23.744840658)

 - Block number: 3 starting at 2021-10-04T07:16:09.379.097.725 (XO: 31.379099476 - 31.379405477, t=37.2 degC)  Rank: 00074
   TM values: Max =   180  Min =  -235  Ave =    -4.37  Std =    27.95
   Min TM value:-235 at an offset of 109.125 usec -> 2021-10-04T07:16:09.379.206.850 (XO: 31.379208601)
   Max TM value:180 at an offset of 111.675 usec -> 2021-10-04T07:16:09.379.209.400 (XO: 31.379211151)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 111.513 usec -> 2021-10-04T07:16:09.379.209.237 (XO: 31.379210989)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 168.788 usec -> 2021-10-04T07:16:09.379.266.513 (XO: 31.379268264)

 - Block number: 4 starting at 2021-10-04T07:16:32.308.965.725 (XO: 54.308968175 - 54.309274177, t=37.2 degC)  Rank: 00099
   TM values: Max =   239  Min =  -270  Ave =    -4.51  Std =    33.89
   Min TM value:-270 at an offset of 81.725 usec -> 2021-10-04T07:16:32.309.047.450 (XO: 54.309049901)
   Max TM value:239 at an offset of 152.826 usec -> 2021-10-04T07:16:32.309.118.550 (XO: 54.309121001)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 125.763 usec -> 2021-10-04T07:16:32.309.091.487 (XO: 54.309093938)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 110.188 usec -> 2021-10-04T07:16:32.309.075.912 (XO: 54.309078363)

 - Block number: 5 starting at 2021-10-04T07:16:38.980.986.725 (XO: 60.980990462 - 60.981296463, t=37.2 degC)  Rank: 00075
   TM values: Max =   176  Min =  -233  Ave =    -4.34  Std =    17.38
   Min TM value:-233 at an offset of 107.013 usec -> 2021-10-04T07:16:38.981.093.737 (XO: 60.981097475)
   Max TM value:176 at an offset of 109.625 usec -> 2021-10-04T07:16:38.981.096.350 (XO: 60.981100088)
   Max dTM/dt [1/us]:1.48e+03 at an offset of 109.400 usec -> 2021-10-04T07:16:38.981.096.125 (XO: 60.981099863)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 106.788 usec -> 2021-10-04T07:16:38.981.093.512 (XO: 60.981097250)

 - Block number: 6 starting at 2021-10-04T07:17:05.703.396.725 (XO: 87.703400046 - 87.703706047, t=37.2 degC)  Rank: 00077
   TM values: Max =   188  Min =  -234  Ave =    -4.52  Std =    20.34
   Min TM value:-234 at an offset of 117.225 usec -> 2021-10-04T07:17:05.703.513.950 (XO: 87.703517272)
   Max TM value:188 at an offset of 119.738 usec -> 2021-10-04T07:17:05.703.516.462 (XO: 87.703519784)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 119.588 usec -> 2021-10-04T07:17:05.703.516.312 (XO: 87.703519634)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 117.013 usec -> 2021-10-04T07:17:05.703.513.737 (XO: 87.703517059)
