-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 29 02:05:21 2022 UTC
-- File 20211004_001317_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.617815 to 68.348092
-each separately from 14s to 68s and from 38055us to 618104us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.459917 in 1 packets: 18
      9.959745 in 1 packets: 36
      18.289643 in 1 packets: 54
      12.019527 in 1 packets: 72
MIU packet time from 2021-10-04T00:13:34.617.815.000 to 2021-10-04T00:14:28.348.092.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.459917 in 1 packets: 18
      9.959745 in 1 packets: 36
      18.289643 in 1 packets: 54
      12.019527 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4402922 to 69.4401125
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999534 in 1 packets: 18
      9.9999667 in 1 packets: 36
      17.9999401 in 1 packets: 54
      11.9999601 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20211004_001317_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.0581081 to 16.0583971
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4598710  -- OK for SL --
   MIU time from 2021-10-04T00:13:34.617.815.000 to 2021-10-04T00:13:34.618.104.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.459917

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5182681 to 29.5185571
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9597110  -- OK for SL --
   MIU time from 2021-10-04T00:13:48.078.021.000 to 2021-10-04T00:13:48.078.310.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.959745

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.4782681 to 39.4785571
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2895830  -- OK for SL --
   MIU time from 2021-10-04T00:13:58.038.055.000 to 2021-10-04T00:13:58.038.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.289643

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.7681401 to 57.7684291
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0194870  -- OK for SL --
   MIU time from 2021-10-04T00:14:16.327.987.000 to 2021-10-04T00:14:16.328.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.019527

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.7879161 to 69.7882051
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-04T00:14:28.347.803.000 to 2021-10-04T00:14:28.348.092.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.058108100 corrected by 40bit 10MHz counter of 15.440292200 obtained at 1Hz counter of 14. Result: 14.617817939
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-04T00:13:34.617.674.725
   Max 1st sample MIU time among all SL packets: 2021-10-04T00:14:28.347.951.725
   Min 1st sample XO time among all SL packets: 14.617677663
   Max 1st sample XO time among all SL packets: 68.347953508
   TM values from -263 to 213 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-04T00:13:34.617.674.725 (XO: 14.617677663 - 14.617983664, t=37.2 degC)  Rank: 00079
   TM values: Max =   206  Min =  -249  Ave =    -4.58  Std =    18.85
   Min TM value:-249 at an offset of 123.650 usec -> 2021-10-04T00:13:34.617.798.375 (XO: 14.617801314)
   Max TM value:206 at an offset of 126.288 usec -> 2021-10-04T00:13:34.617.801.012 (XO: 14.617803951)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 126.125 usec -> 2021-10-04T00:13:34.617.800.850 (XO: 14.617803789)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 123.488 usec -> 2021-10-04T00:13:34.617.798.212 (XO: 14.617801151)

 - Block number: 2 starting at 2021-10-04T00:13:48.077.880.725 (XO: 28.077882490 - 28.078188491, t=37.2 degC)  Rank: 00079
   TM values: Max =   191  Min =  -238  Ave =    -4.39  Std =    18.01
   Min TM value:-238 at an offset of 125.075 usec -> 2021-10-04T00:13:48.078.005.800 (XO: 28.078007565)
   Max TM value:191 at an offset of 124.800 usec -> 2021-10-04T00:13:48.078.005.525 (XO: 28.078007290)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 122.650 usec -> 2021-10-04T00:13:48.078.003.375 (XO: 28.078005140)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 124.913 usec -> 2021-10-04T00:13:48.078.005.637 (XO: 28.078007403)

 - Block number: 3 starting at 2021-10-04T00:13:58.037.914.725 (XO: 38.037915650 - 38.038221651, t=37.2 degC)  Rank: 00078
   TM values: Max =   186  Min =  -249  Ave =    -4.57  Std =    17.87
   Min TM value:-249 at an offset of 128.063 usec -> 2021-10-04T00:13:58.038.042.787 (XO: 38.038043713)
   Max TM value:186 at an offset of 127.788 usec -> 2021-10-04T00:13:58.038.042.512 (XO: 38.038043438)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 130.525 usec -> 2021-10-04T00:13:58.038.045.250 (XO: 38.038046176)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 127.913 usec -> 2021-10-04T00:13:58.038.042.637 (XO: 38.038043563)

 - Block number: 4 starting at 2021-10-04T00:14:16.327.846.725 (XO: 56.327848507 - 56.328154508, t=37.2 degC)  Rank: 00087
   TM values: Max =   213  Min =  -237  Ave =    -4.43  Std =    17.53
   Min TM value:-237 at an offset of 125.488 usec -> 2021-10-04T00:14:16.327.972.212 (XO: 56.327973995)
   Max TM value:213 at an offset of 128.013 usec -> 2021-10-04T00:14:16.327.974.737 (XO: 56.327976520)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 127.863 usec -> 2021-10-04T00:14:16.327.974.587 (XO: 56.327976370)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 125.313 usec -> 2021-10-04T00:14:16.327.972.037 (XO: 56.327973820)

 - Block number: 5 starting at 2021-10-04T00:14:28.347.662.725 (XO: 68.347664507 - 68.347970508, t=37.2 degC)  Rank: 00075
   TM values: Max =   197  Min =  -263  Ave =    -4.49  Std =    18.89
   Min TM value:-263 at an offset of 125.988 usec -> 2021-10-04T00:14:28.347.788.712 (XO: 68.347790495)
   Max TM value:197 at an offset of 125.688 usec -> 2021-10-04T00:14:28.347.788.412 (XO: 68.347790195)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 123.513 usec -> 2021-10-04T00:14:28.347.786.237 (XO: 68.347788020)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 125.788 usec -> 2021-10-04T00:14:28.347.788.512 (XO: 68.347790295)
