-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 28 06:21:43 2022 UTC
-- File 20211003_094556_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.268909 to 62.369580
-each separately from 10s to 62s and from 268909us to 966090us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.696603 in 1 packets: 18
      20.575941 in 1 packets: 36
      12.787674 in 1 packets: 54
      5.039008 in 1 packets: 72
MIU packet time from 2021-10-03T09:46:09.268.909.000 to 2021-10-03T09:47:01.369.580.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.696603 in 1 packets: 18
      20.575941 in 1 packets: 36
      12.787674 in 1 packets: 54
      5.039008 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8440875 to 63.8439132
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999564 in 1 packets: 18
      20.9999296 in 1 packets: 36
      12.9999565 in 1 packets: 54
      4.9999832 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 90 packets (all packets)
Number of generated alerts: 
      0 in 90 packets (all packets)
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20211003_094556_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1129974 to 12.1132864
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6965590  -- OK for SL --
   MIU time from 2021-10-03T09:46:09.268.909.000 to 2021-10-03T09:46:09.269.198.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.696603

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.8098454 to 25.8101344
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5758710  -- OK for SL --
   MIU time from 2021-10-03T09:46:22.965.801.000 to 2021-10-03T09:46:22.966.090.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.575941

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.3860054 to 46.3862944
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.7876310  -- OK for SL --
   MIU time from 2021-10-03T09:46:43.542.031.000 to 2021-10-03T09:46:43.542.320.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.787674

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1739254 to 59.1742144
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.0389910  -- OK for SL --
   MIU time from 2021-10-03T09:46:56.329.994.000 to 2021-10-03T09:46:56.330.283.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.039008

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.2132054 to 64.2134944
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-03T09:47:01.369.291.000 to 2021-10-03T09:47:01.369.580.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.112997400 corrected by 40bit 10MHz counter of 11.844087500 obtained at 1Hz counter of 10. Result: 10.268910814
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-03T09:46:09.268.768.725
   Max 1st sample MIU time among all SL packets: 2021-10-03T09:47:01.369.439.725
   Min 1st sample XO time among all SL packets: 10.268770539
   Max 1st sample XO time among all SL packets: 62.369442181
   TM values from -260 to 229 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-03T09:46:09.268.768.725 (XO: 10.268770539 - 10.269076540, t=37.2 degC)  Rank: 00078
   TM values: Max =   210  Min =  -246  Ave =    -4.49  Std =    21.75
   Min TM value:-246 at an offset of 101.738 usec -> 2021-10-03T09:46:09.268.870.462 (XO: 10.268872277)
   Max TM value:210 at an offset of 104.238 usec -> 2021-10-03T09:46:09.268.872.962 (XO: 10.268874777)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 104.100 usec -> 2021-10-03T09:46:09.268.872.825 (XO: 10.268874639)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 101.563 usec -> 2021-10-03T09:46:09.268.870.287 (XO: 10.268872102)

 - Block number: 2 starting at 2021-10-03T09:46:22.965.660.725 (XO: 23.965664412 - 23.965970413, t=37.2 degC)  Rank: 00080
   TM values: Max =   229  Min =  -246  Ave =    -4.47  Std =    18.96
   Min TM value:-246 at an offset of 129.563 usec -> 2021-10-03T09:46:22.965.790.287 (XO: 23.965793975)
   Max TM value:229 at an offset of 129.300 usec -> 2021-10-03T09:46:22.965.790.025 (XO: 23.965793712)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 131.950 usec -> 2021-10-03T09:46:22.965.792.675 (XO: 23.965796362)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 129.425 usec -> 2021-10-03T09:46:22.965.790.150 (XO: 23.965793837)

 - Block number: 3 starting at 2021-10-03T09:46:43.541.890.725 (XO: 44.541893413 - 44.542199414, t=37.2 degC)  Rank: 00076
   TM values: Max =   187  Min =  -260  Ave =    -4.57  Std =    22.20
   Min TM value:-260 at an offset of 125.875 usec -> 2021-10-03T09:46:43.542.016.600 (XO: 44.542019289)
   Max TM value:187 at an offset of 128.438 usec -> 2021-10-03T09:46:43.542.019.162 (XO: 44.542021851)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 128.300 usec -> 2021-10-03T09:46:43.542.019.025 (XO: 44.542021714)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 125.750 usec -> 2021-10-03T09:46:43.542.016.475 (XO: 44.542019164)

 - Block number: 4 starting at 2021-10-03T09:46:56.329.853.725 (XO: 57.329856247 - 57.330162248, t=37.2 degC)  Rank: 00070
   TM values: Max =   196  Min =  -233  Ave =    -4.44  Std =    25.63
   Min TM value:-233 at an offset of 127.488 usec -> 2021-10-03T09:46:56.329.981.212 (XO: 57.329983734)
   Max TM value:196 at an offset of 130.150 usec -> 2021-10-03T09:46:56.329.983.875 (XO: 57.329986397)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 185.401 usec -> 2021-10-03T09:46:56.330.039.125 (XO: 57.330041647)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 127.350 usec -> 2021-10-03T09:46:56.329.981.075 (XO: 57.329983597)

 - Block number: 5 starting at 2021-10-03T09:47:01.369.150.725 (XO: 62.369153180 - 62.369459181, t=37.2 degC)  Rank: 00076
   TM values: Max =   190  Min =  -240  Ave =    -4.63  Std =    20.72
   Min TM value:-240 at an offset of 115.950 usec -> 2021-10-03T09:47:01.369.266.675 (XO: 62.369269131)
   Max TM value:190 at an offset of 118.588 usec -> 2021-10-03T09:47:01.369.269.312 (XO: 62.369271768)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 118.438 usec -> 2021-10-03T09:47:01.369.269.162 (XO: 62.369271618)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 115.813 usec -> 2021-10-03T09:47:01.369.266.537 (XO: 62.369268993)
