-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 27 22:36:08 2022 UTC
-- File 20211003_034943_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.269830 to 19.250264
-each separately from 9s to 19s and from 249975us to 270119us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.979856 in 1 packets: 18
MIU packet time from 2021-10-03T03:49:55.269.830.000 to 2021-10-03T03:50:05.250.264.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.979856 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8682532 to 20.8682198
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999666 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211003_034943_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1380842 to 11.1383732
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9798230  -- OK for SL --
   MIU time from 2021-10-03T03:49:55.269.830.000 to 2021-10-03T03:49:55.270.119.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.979856

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1181962 to 21.1184852
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-03T03:50:05.249.975.000 to 2021-10-03T03:50:05.250.264.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.138084200 corrected by 40bit 10MHz counter of 10.868253200 obtained at 1Hz counter of 9. Result: 9.269831917
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-03T03:49:55.269.689.725
   Max 1st sample MIU time among all SL packets: 2021-10-03T03:50:05.250.123.725
   Min 1st sample XO time among all SL packets: 9.269691642
   Max 1st sample XO time among all SL packets: 19.250125975
   TM values from -258 to 211 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-03T03:49:55.269.689.725 (XO: 9.269691642 - 9.269997643, t=37.2 degC)  Rank: 00081
   TM values: Max =   211  Min =  -258  Ave =    -4.51  Std =    31.72
   Min TM value:-258 at an offset of 113.613 usec -> 2021-10-03T03:49:55.269.803.337 (XO: 9.269805255)
   Max TM value:211 at an offset of 113.313 usec -> 2021-10-03T03:49:55.269.803.037 (XO: 9.269804955)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 116.000 usec -> 2021-10-03T03:49:55.269.805.725 (XO: 9.269807642)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 113.438 usec -> 2021-10-03T03:49:55.269.803.162 (XO: 9.269805080)

 - Block number: 2 starting at 2021-10-03T03:50:05.249.834.725 (XO: 19.249836974 - 19.250142975, t=37.2 degC)  Rank: 00086
   TM values: Max =   206  Min =  -258  Ave =    -4.48  Std =    21.43
   Min TM value:-258 at an offset of 123.025 usec -> 2021-10-03T03:50:05.249.957.750 (XO: 19.249960000)
   Max TM value:206 at an offset of 125.600 usec -> 2021-10-03T03:50:05.249.960.325 (XO: 19.249962575)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 125.425 usec -> 2021-10-03T03:50:05.249.960.150 (XO: 19.249962400)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 122.850 usec -> 2021-10-03T03:50:05.249.957.575 (XO: 19.249959825)
