-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 27 09:45:36 2022 UTC
-- File 20211002_223846_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.315524 to 81.442304
-each separately from 6s to 81s and from 39391us to 796283us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.480181 in 1 packets: 18
      7.339705 in 1 packets: 36
      22.091497 in 1 packets: 54
      9.811328 in 1 packets: 72
      19.402335 in 1 packets: 90
MIU packet time from 2021-10-02T22:38:55.315.524.000 to 2021-10-02T22:40:10.442.304.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.480181 in 1 packets: 18
      7.339705 in 1 packets: 36
      22.091497 in 1 packets: 54
      9.811328 in 1 packets: 72
      19.402335 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8983260 to 82.8980748
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999464 in 1 packets: 18
      7.9999733 in 1 packets: 36
      21.9999263 in 1 packets: 54
      9.9999666 in 1 packets: 72
      18.9999362 in 1 packets: 90
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_223846_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2138510 to 8.2141400
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4801270  -- OK for SL --
   MIU time from 2021-10-02T22:38:55.315.524.000 to 2021-10-02T22:38:55.315.813.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.480181

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6942670 to 24.6945560
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.3396790  -- OK for SL --
   MIU time from 2021-10-02T22:39:11.795.994.000 to 2021-10-02T22:39:11.796.283.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.339705

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0342350 to 32.0345240
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.0914230  -- OK for SL --
   MIU time from 2021-10-02T22:39:19.135.988.000 to 2021-10-02T22:39:19.136.277.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.091497

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.1259470 to 54.1262360
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.8112950  -- OK for SL --
   MIU time from 2021-10-02T22:39:41.227.774.000 to 2021-10-02T22:39:41.228.063.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.811328

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.9375310 to 63.9378200
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4022710  -- OK for SL --
   MIU time from 2021-10-02T22:39:51.039.391.000 to 2021-10-02T22:39:51.039.680.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.402335

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 83.3400910 to 83.3403800
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T22:40:10.442.015.000 to 2021-10-02T22:40:10.442.304.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.213851000 corrected by 40bit 10MHz counter of 7.898326000 obtained at 1Hz counter of 6. Result: 6.315526041
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T22:38:55.315.383.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T22:40:10.442.163.725
   Min 1st sample XO time among all SL packets: 6.315385766
   Max 1st sample XO time among all SL packets: 81.442166384
   TM values from -309 to 265 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T22:38:55.315.383.725 (XO: 6.315385766 - 6.315691767, t=37.2 degC)  Rank: 00079
   TM values: Max =   220  Min =  -253  Ave =    -4.56  Std =    19.69
   Min TM value:-253 at an offset of 124.738 usec -> 2021-10-02T22:38:55.315.508.462 (XO: 6.315510504)
   Max TM value:220 at an offset of 127.350 usec -> 2021-10-02T22:38:55.315.511.075 (XO: 6.315513116)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 127.225 usec -> 2021-10-02T22:38:55.315.510.950 (XO: 6.315512991)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 124.525 usec -> 2021-10-02T22:38:55.315.508.250 (XO: 6.315510291)

 - Block number: 2 starting at 2021-10-02T22:39:11.795.853.725 (XO: 22.795856951 - 22.796162952, t=37.2 degC)  Rank: 00079
   TM values: Max =   212  Min =  -225  Ave =    -4.46  Std =    18.48
   Min TM value:-225 at an offset of 123.875 usec -> 2021-10-02T22:39:11.795.977.600 (XO: 22.795980827)
   Max TM value:212 at an offset of 126.563 usec -> 2021-10-02T22:39:11.795.980.287 (XO: 22.795983514)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 126.475 usec -> 2021-10-02T22:39:11.795.980.200 (XO: 22.795983427)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 123.700 usec -> 2021-10-02T22:39:11.795.977.425 (XO: 22.795980652)

 - Block number: 3 starting at 2021-10-02T22:39:19.135.847.725 (XO: 30.135849487 - 30.136155488, t=37.2 degC)  Rank: 00078
   TM values: Max =   256  Min =  -309  Ave =    -4.49  Std =    21.99
   Min TM value:-309 at an offset of 112.288 usec -> 2021-10-02T22:39:19.135.960.012 (XO: 30.135961775)
   Max TM value:256 at an offset of 112.038 usec -> 2021-10-02T22:39:19.135.959.762 (XO: 30.135961525)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 114.688 usec -> 2021-10-02T22:39:19.135.962.412 (XO: 30.135964175)
   Min dTM/dt [1/us]:-3.17e+03 at an offset of 112.163 usec -> 2021-10-02T22:39:19.135.959.887 (XO: 30.135961650)

 - Block number: 4 starting at 2021-10-02T22:39:41.227.633.725 (XO: 52.227635476 - 52.227941477, t=37.2 degC)  Rank: 00083
   TM values: Max =   265  Min =  -283  Ave =    -4.47  Std =    21.26
   Min TM value:-283 at an offset of 117.013 usec -> 2021-10-02T22:39:41.227.750.737 (XO: 52.227752489)
   Max TM value:265 at an offset of 114.913 usec -> 2021-10-02T22:39:41.227.748.637 (XO: 52.227750389)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 119.400 usec -> 2021-10-02T22:39:41.227.753.125 (XO: 52.227754877)
   Min dTM/dt [1/us]:-2.70e+03 at an offset of 116.875 usec -> 2021-10-02T22:39:41.227.750.600 (XO: 52.227752352)

 - Block number: 5 starting at 2021-10-02T22:39:51.039.250.725 (XO: 62.039252258 - 62.039558259, t=37.2 degC)  Rank: 00082
   TM values: Max =   213  Min =  -267  Ave =    -4.48  Std =    23.73
   Min TM value:-267 at an offset of 123.763 usec -> 2021-10-02T22:39:51.039.374.487 (XO: 62.039376021)
   Max TM value:213 at an offset of 123.525 usec -> 2021-10-02T22:39:51.039.374.250 (XO: 62.039375784)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 126.163 usec -> 2021-10-02T22:39:51.039.376.887 (XO: 62.039378421)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 123.613 usec -> 2021-10-02T22:39:51.039.374.337 (XO: 62.039375871)

 - Block number: 6 starting at 2021-10-02T22:40:10.441.874.725 (XO: 81.441877383 - 81.442183384, t=37.2 degC)  Rank: 00080
   TM values: Max =   240  Min =  -273  Ave =    -4.46  Std =    27.41
   Min TM value:-273 at an offset of 112.775 usec -> 2021-10-02T22:40:10.441.987.500 (XO: 81.441990159)
   Max TM value:240 at an offset of 115.388 usec -> 2021-10-02T22:40:10.441.990.112 (XO: 81.441992771)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 115.188 usec -> 2021-10-02T22:40:10.441.989.912 (XO: 81.441992571)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 112.563 usec -> 2021-10-02T22:40:10.441.987.287 (XO: 81.441989946)
