-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 27 09:00:11 2022 UTC
-- File 20211002_220909_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.787599 to 57.237789
-each separately from 10s to 57s and from 237500us to 787888us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.649680 in 1 packets: 18
      18.279851 in 1 packets: 36
      23.519503 in 1 packets: 54
MIU packet time from 2021-10-02T22:09:22.787.599.000 to 2021-10-02T22:10:09.237.789.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.649680 in 1 packets: 18
      18.279851 in 1 packets: 36
      23.519503 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8872879 to 58.8871308
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999833 in 1 packets: 18
      17.9999399 in 1 packets: 36
      23.9999197 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_220909_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.6748880 to 12.6751770
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.6496630  -- OK for SL --
   MIU time from 2021-10-02T22:09:22.787.599.000 to 2021-10-02T22:09:22.787.888.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.649680

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3248400 to 17.3251290
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2797910  -- OK for SL --
   MIU time from 2021-10-02T22:09:27.437.568.000 to 2021-10-02T22:09:27.437.857.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.279851

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6049200 to 35.6052090
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5194230  -- OK for SL --
   MIU time from 2021-10-02T22:09:45.717.708.000 to 2021-10-02T22:09:45.717.997.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.519503

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1246320 to 59.1249210
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T22:10:09.237.500.000 to 2021-10-02T22:10:09.237.789.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.674888000 corrected by 40bit 10MHz counter of 11.887287900 obtained at 1Hz counter of 10. Result: 10.787602699
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T22:09:22.787.458.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T22:10:09.237.648.725
   Min 1st sample XO time among all SL packets: 10.787462424
   Max 1st sample XO time among all SL packets: 57.237650709
   TM values from -261 to 250 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T22:09:22.787.458.725 (XO: 10.787462424 - 10.787768425, t=37.2 degC)  Rank: 00081
   TM values: Max =   250  Min =  -257  Ave =    -4.56  Std =    26.89
   Min TM value:-257 at an offset of 123.413 usec -> 2021-10-02T22:09:22.787.582.137 (XO: 10.787585837)
   Max TM value:250 at an offset of 126.013 usec -> 2021-10-02T22:09:22.787.584.737 (XO: 10.787588437)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 125.875 usec -> 2021-10-02T22:09:22.787.584.600 (XO: 10.787588299)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 123.225 usec -> 2021-10-02T22:09:22.787.581.950 (XO: 10.787585649)

 - Block number: 2 starting at 2021-10-02T22:09:27.437.427.725 (XO: 15.437429969 - 15.437735970, t=37.2 degC)  Rank: 00079
   TM values: Max =   224  Min =  -248  Ave =    -4.52  Std =    29.10
   Min TM value:-248 at an offset of 111.688 usec -> 2021-10-02T22:09:27.437.539.412 (XO: 15.437541656)
   Max TM value:224 at an offset of 114.463 usec -> 2021-10-02T22:09:27.437.542.187 (XO: 15.437544431)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 114.250 usec -> 2021-10-02T22:09:27.437.541.975 (XO: 15.437544219)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 111.488 usec -> 2021-10-02T22:09:27.437.539.212 (XO: 15.437541456)

 - Block number: 3 starting at 2021-10-02T22:09:45.717.567.725 (XO: 33.717571065 - 33.717877066, t=37.2 degC)  Rank: 00078
   TM values: Max =   231  Min =  -261  Ave =    -4.48  Std =    22.28
   Min TM value:-261 at an offset of 97.613 usec -> 2021-10-02T22:09:45.717.665.337 (XO: 33.717668678)
   Max TM value:231 at an offset of 112.925 usec -> 2021-10-02T22:09:45.717.680.650 (XO: 33.717683990)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 112.800 usec -> 2021-10-02T22:09:45.717.680.525 (XO: 33.717683865)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 110.100 usec -> 2021-10-02T22:09:45.717.677.825 (XO: 33.717681165)

 - Block number: 4 starting at 2021-10-02T22:10:09.237.359.725 (XO: 57.237361708 - 57.237667709, t=37.2 degC)  Rank: 00083
   TM values: Max =   228  Min =  -260  Ave =    -4.46  Std =    22.77
   Min TM value:-260 at an offset of 125.800 usec -> 2021-10-02T22:10:09.237.485.525 (XO: 57.237487509)
   Max TM value:228 at an offset of 126.075 usec -> 2021-10-02T22:10:09.237.485.800 (XO: 57.237487784)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 125.875 usec -> 2021-10-02T22:10:09.237.485.600 (XO: 57.237487584)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 123.313 usec -> 2021-10-02T22:10:09.237.483.037 (XO: 57.237485021)
