-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 27 08:17:18 2022 UTC
-- File 20211002_214133_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.743150 to 39.590621
-each separately from 6s to 39s and from 495654us to 743439us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.752215 in 1 packets: 18
      21.094389 in 1 packets: 36
MIU packet time from 2021-10-02T21:41:42.743.150.000 to 2021-10-02T21:42:15.590.621.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.752215 in 1 packets: 18
      21.094389 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8961145 to 40.8960045
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999600 in 1 packets: 18
      20.9999300 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_214133_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6392651 to 8.6395541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7521750  -- OK for SL --
   MIU time from 2021-10-02T21:41:42.743.150.000 to 2021-10-02T21:41:42.743.439.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.752215

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3917291 to 20.3920181
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0943190  -- OK for SL --
   MIU time from 2021-10-02T21:41:54.495.654.000 to 2021-10-02T21:41:54.495.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.094389

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.4863371 to 41.4866261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T21:42:15.590.332.000 to 2021-10-02T21:42:15.590.621.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.639265100 corrected by 40bit 10MHz counter of 7.896114500 obtained at 1Hz counter of 6. Result: 6.743153052
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T21:41:42.743.009.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T21:42:15.590.480.725
   Min 1st sample XO time among all SL packets: 6.743012777
   Max 1st sample XO time among all SL packets: 39.590483274
   TM values from -287 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T21:41:42.743.009.725 (XO: 6.743012777 - 6.743318778, t=37.2 degC)  Rank: 00082
   TM values: Max =   225  Min =  -287  Ave =    -4.46  Std =    30.78
   Min TM value:-287 at an offset of 183.126 usec -> 2021-10-02T21:41:42.743.192.850 (XO: 6.743195903)
   Max TM value:225 at an offset of 126.200 usec -> 2021-10-02T21:41:42.743.135.925 (XO: 6.743138977)
   Max dTM/dt [1/us]:2.31e+03 at an offset of 128.900 usec -> 2021-10-02T21:41:42.743.138.625 (XO: 6.743141677)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 126.338 usec -> 2021-10-02T21:41:42.743.136.062 (XO: 6.743139115)

 - Block number: 2 starting at 2021-10-02T21:41:54.495.513.725 (XO: 18.495515960 - 18.495821961, t=37.2 degC)  Rank: 00078
   TM values: Max =   222  Min =  -272  Ave =    -4.50  Std =    23.84
   Min TM value:-272 at an offset of 121.713 usec -> 2021-10-02T21:41:54.495.635.437 (XO: 18.495637673)
   Max TM value:222 at an offset of 124.288 usec -> 2021-10-02T21:41:54.495.638.012 (XO: 18.495640248)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 124.125 usec -> 2021-10-02T21:41:54.495.637.850 (XO: 18.495640086)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 121.500 usec -> 2021-10-02T21:41:54.495.635.225 (XO: 18.495637461)

 - Block number: 3 starting at 2021-10-02T21:42:15.590.191.725 (XO: 39.590194273 - 39.590500274, t=37.2 degC)  Rank: 00088
   TM values: Max =   224  Min =  -218  Ave =    -4.43  Std =    20.92
   Min TM value:-218 at an offset of 121.938 usec -> 2021-10-02T21:42:15.590.313.662 (XO: 39.590316211)
   Max TM value:224 at an offset of 124.700 usec -> 2021-10-02T21:42:15.590.316.425 (XO: 39.590318973)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 124.563 usec -> 2021-10-02T21:42:15.590.316.287 (XO: 39.590318836)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 109.463 usec -> 2021-10-02T21:42:15.590.301.187 (XO: 39.590303736)
