-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 26 23:16:37 2022 UTC
-- File 20211002_104249_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.525547 to 21.201426
-each separately from 0s to 21s and from 201137us to 525836us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.675301 in 1 packets: 18
MIU packet time from 2021-10-02T10:42:52.525.547.000 to 2021-10-02T10:43:13.201.426.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.675301 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9284931 to 22.9284230
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999299 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_104249_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.4540413 to 2.4543303
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.6752310  -- OK for SL --
   MIU time from 2021-10-02T10:42:52.525.547.000 to 2021-10-02T10:42:52.525.836.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.675301

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1295613 to 23.1298503
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T10:43:13.201.137.000 to 2021-10-02T10:43:13.201.426.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.454041300 corrected by 40bit 10MHz counter of 1.928493100 obtained at 1Hz counter of 0. Result: 0.525549934
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T10:42:52.525.406.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T10:43:13.201.285.725
   Min 1st sample XO time among all SL packets: 0.525409659
   Max 1st sample XO time among all SL packets: 21.201287689
   TM values from -267 to 244 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T10:42:52.525.406.725 (XO: 0.525409659 - 0.525715660, t=37.2 degC)  Rank: 00080
   TM values: Max =   207  Min =  -237  Ave =    -4.41  Std =    21.08
   Min TM value:-237 at an offset of 89.950 usec -> 2021-10-02T10:42:52.525.496.675 (XO: 0.525499609)
   Max TM value:207 at an offset of 92.550 usec -> 2021-10-02T10:42:52.525.499.275 (XO: 0.525502209)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 92.400 usec -> 2021-10-02T10:42:52.525.499.125 (XO: 0.525502059)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 89.738 usec -> 2021-10-02T10:42:52.525.496.462 (XO: 0.525499397)

 - Block number: 2 starting at 2021-10-02T10:43:13.200.996.725 (XO: 21.200998688 - 21.201304689, t=37.2 degC)  Rank: 00091
   TM values: Max =   244  Min =  -267  Ave =    -4.55  Std =    17.75
   Min TM value:-267 at an offset of 129.513 usec -> 2021-10-02T10:43:13.201.126.237 (XO: 21.201128201)
   Max TM value:244 at an offset of 127.400 usec -> 2021-10-02T10:43:13.201.124.125 (XO: 21.201126089)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 127.263 usec -> 2021-10-02T10:43:13.201.123.987 (XO: 21.201125951)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 129.375 usec -> 2021-10-02T10:43:13.201.126.100 (XO: 21.201128064)
