-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 26 11:59:23 2022 UTC
-- File 20211002_092034_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.668895 to 20.629118
-each separately from 6s to 20s and from 628829us to 669184us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.959645 in 1 packets: 18
MIU packet time from 2021-10-02T09:20:43.668.895.000 to 2021-10-02T09:20:57.629.118.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.959645 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4532193 to 21.4531728
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999535 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_092034_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.1221148 to 8.1224038
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.9595990  -- OK for SL --
   MIU time from 2021-10-02T09:20:43.668.895.000 to 2021-10-02T09:20:43.669.184.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.959645

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0820028 to 22.0822918
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T09:20:57.628.829.000 to 2021-10-02T09:20:57.629.118.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.122114800 corrected by 40bit 10MHz counter of 7.453219300 obtained at 1Hz counter of 6. Result: 6.668897707
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T09:20:43.668.754.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T09:20:57.628.977.725
   Min 1st sample XO time among all SL packets: 6.668757432
   Max 1st sample XO time among all SL packets: 20.628980801
   TM values from -217 to 185 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T09:20:43.668.754.725 (XO: 6.668757432 - 6.669063433, t=37.2 degC)  Rank: 00082
   TM values: Max =   176  Min =  -205  Ave =    -4.48  Std =    21.31
   Min TM value:-205 at an offset of 119.763 usec -> 2021-10-02T09:20:43.668.874.487 (XO: 6.668877195)
   Max TM value:176 at an offset of 122.500 usec -> 2021-10-02T09:20:43.668.877.225 (XO: 6.668879932)
   Max dTM/dt [1/us]:1.35e+03 at an offset of 122.350 usec -> 2021-10-02T09:20:43.668.877.075 (XO: 6.668879782)
   Min dTM/dt [1/us]:-1.33e+03 at an offset of 119.500 usec -> 2021-10-02T09:20:43.668.874.225 (XO: 6.668876932)

 - Block number: 2 starting at 2021-10-02T09:20:57.628.688.725 (XO: 20.628691800 - 20.628997801, t=37.2 degC)  Rank: 00079
   TM values: Max =   185  Min =  -217  Ave =    -4.46  Std =    16.55
   Min TM value:-217 at an offset of 108.938 usec -> 2021-10-02T09:20:57.628.797.662 (XO: 20.628800738)
   Max TM value:185 at an offset of 111.563 usec -> 2021-10-02T09:20:57.628.800.287 (XO: 20.628803363)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 111.375 usec -> 2021-10-02T09:20:57.628.800.100 (XO: 20.628803175)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 108.763 usec -> 2021-10-02T09:20:57.628.797.487 (XO: 20.628800563)
