-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 26 09:06:50 2022 UTC
-- File 20211002_013425_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.683063 to 42.243636
-each separately from 5s to 42s and from 243347us to 683352us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.609890 in 1 packets: 18
      16.949816 in 1 packets: 36
MIU packet time from 2021-10-02T01:34:33.683.063.000 to 2021-10-02T01:35:10.243.636.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.609890 in 1 packets: 18
      16.949816 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9048680 to 43.9047446
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999333 in 1 packets: 18
      16.9999433 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20211002_013425_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.5879322 to 7.5882212
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6098230  -- OK for SL --
   MIU time from 2021-10-02T01:34:33.683.063.000 to 2021-10-02T01:34:33.683.352.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.609890

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.1980442 to 27.1983332
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9497590  -- OK for SL --
   MIU time from 2021-10-02T01:34:53.293.242.000 to 2021-10-02T01:34:53.293.531.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.949816

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.1480922 to 44.1483812
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-02T01:35:10.243.347.000 to 2021-10-02T01:35:10.243.636.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.587932200 corrected by 40bit 10MHz counter of 6.904868000 obtained at 1Hz counter of 5. Result: 5.683066454
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-02T01:34:33.682.922.725
   Max 1st sample MIU time among all SL packets: 2021-10-02T01:35:10.243.495.725
   Min 1st sample XO time among all SL packets: 5.682926179
   Max 1st sample XO time among all SL packets: 42.243497153
   TM values from -258 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-02T01:34:33.682.922.725 (XO: 5.682926179 - 5.683232180, t=37.2 degC)  Rank: 00087
   TM values: Max =   230  Min =  -256  Ave =    -4.40  Std =    23.48
   Min TM value:-256 at an offset of 156.126 usec -> 2021-10-02T01:34:33.683.078.850 (XO: 5.683082304)
   Max TM value:230 at an offset of 131.050 usec -> 2021-10-02T01:34:33.683.053.775 (XO: 5.683057229)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 126.213 usec -> 2021-10-02T01:34:33.683.048.937 (XO: 5.683052392)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 128.350 usec -> 2021-10-02T01:34:33.683.051.075 (XO: 5.683054529)

 - Block number: 2 starting at 2021-10-02T01:34:53.293.101.725 (XO: 25.293103622 - 25.293409623, t=37.2 degC)  Rank: 00079
   TM values: Max =   202  Min =  -245  Ave =    -4.53  Std =    18.89
   Min TM value:-245 at an offset of 126.163 usec -> 2021-10-02T01:34:53.293.227.887 (XO: 25.293229784)
   Max TM value:202 at an offset of 128.788 usec -> 2021-10-02T01:34:53.293.230.512 (XO: 25.293232409)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 128.638 usec -> 2021-10-02T01:34:53.293.230.362 (XO: 25.293232259)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 125.950 usec -> 2021-10-02T01:34:53.293.227.675 (XO: 25.293229572)

 - Block number: 3 starting at 2021-10-02T01:35:10.243.206.725 (XO: 42.243208152 - 42.243514153, t=37.2 degC)  Rank: 00088
   TM values: Max =   228  Min =  -258  Ave =    -4.39  Std =    19.64
   Min TM value:-258 at an offset of 125.663 usec -> 2021-10-02T01:35:10.243.332.387 (XO: 42.243333815)
   Max TM value:228 at an offset of 125.400 usec -> 2021-10-02T01:35:10.243.332.125 (XO: 42.243333552)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 123.363 usec -> 2021-10-02T01:35:10.243.330.087 (XO: 42.243331515)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 125.500 usec -> 2021-10-02T01:35:10.243.332.225 (XO: 42.243333652)
